In this study, the AuNP k-shell fluorescence spectra were measured by using a 100 mm long collimator to improve the detectability. The AuNPs were suspended in deionized water at different concentrations. The AuNP suspensions were excited by a micro focal spot x-ray tube with 130 kVp, 300 μA x-ray exposures. The emissions of the AuNP fluorescence were measured by a spectrometer located with an angle of 90 degrees with respect to the excitation beam. The fluorescence acquisition durations for each concentration mode were 3000 s. A 1.0 mm Pb filter and a 1 mm Al filter were utilized to optimize excitation beam and fluorescence emission, respectively. As a result, the k-shell fluorescence peaks, 66.99 keV and 68.80 keV of AuNP, were measured and observed in 0.1, 0.2, 0.4, 0.8, 1.0, 2.0, 4.0 mg/mL concentration modes. The linear relationship between the AuNP suspension concentrations and the number of photons of the fluorescence peaks were observed in the range of 0.1–4.0 mg/mL. The results of experimental measurements demonstrated up to 0.1 mg/mL (0.01 % in weight concentration) detectability.
Di Wu, Molly D. Wong, Yuhua Li, Muhammad U. Ghani, Bin Zheng, Wei R. Chen, and Hong Liu, "Measurements of gold nanoparticle concentration with K-shell x-ray fluorescence spectrum," Proc. SPIE 10065, Biophotonics and Immune Responses XII, 100650M (Presented at SPIE BiOS: January 31, 2017; Published: 20 February 2017); https://doi.org/10.1117/12.2257147.
Conference Presentations are recordings of oral presentations given at SPIE conferences and published as part of the proceedings. They include the speaker's narration with video of the slides and animations. Most include full-text papers. Interactive, searchable transcripts and closed captioning are now available for most presentations.
Search our growing collection of more than 18,000 conference presentations, including many plenaries and keynotes.
Global observations from a well-calibrated passive microwave atmospheric sounder on a CubeSat: Temporal Experiment for Storms and Tropical Systems Technology Demonstration (TEMPEST-D) Mission (Conference Presentation)