A miniature Raman spectrometer was designed in a rapid development cycle (< 4 months) to investigate the performance capabilities achievable with two dimensional (2D) CMOS detectors found in cell phone camera modules and commercial off the shelf optics (COTS). This paper examines the design considerations and tradeoffs made during the development cycle. The final system developed measures 40 mm in length, 40 mm in width, 15 mm tall and couples directly with the cell phone camera optics. Two variants were made: one with an excitation wavelength of 638 nm and the other with a 785 nm excitation wavelength. Raman spectra of the following samples were gathered at both excitations: Toluene, Cyclohexane, Bis(MSB), Aspirin, Urea, and Ammonium Nitrate. The system obtained a resolution of 40 cm-1. The spectra produced at 785 nm excitation required integration times of up to 10 times longer than the 1.5 seconds at 638 nm, however, contained reduced stray light and less fluorescence which led to an overall cleaner signal.
Bryan Auz, Joseph Bonvallet, John Rodriguez, and Ty Olmstead, "Miniature Raman spectroscopy utilizing stabilized diode lasers and 2D CMOS detector arrays," Proc. SPIE 10110, Photonic Instrumentation Engineering IV, 1011019 (Presented at SPIE OPTO: February 03, 2017; Published: 20 February 2017); https://doi.org/10.1117/12.2252396.
Conference Presentations are recordings of oral presentations given at SPIE conferences and published as part of the conference proceedings. They include the speaker's narration along with a video recording of the presentation slides and animations. Many conference presentations also include full-text papers. Search and browse our growing collection of more than 12,000 conference presentations, including many plenary and keynote presentations.