From Event: SPIE Optical Engineering + Applications, 2017
X-ray focusing mirrors with elliptical shape are widely used in synchrotron facilities for micro-, nano-scale focusing experiments. Surface interferometry plays an important role in the x-ray mirrors figuring with subnanometer accuracy. To avoid the second order error in stitching interferometry, relative angle determinable stitching interferometry (RADSI) is under development. This method was first developed by Yamauchi et al from Osaka University, which uses a planar mirror to correct the relative stitching angle between the neighboring subapertures. Here, we use RADSI to measure the x-ray spherical and elliptical mirrors with 300mm aperture Fizeau interferometer. The interferometer is combined with 4 accurate rotation and tilt stages for the stitching measurement. To ensure the stitching accuracy, we first studied the measurement accuracy within every single subaperture. Multiple measurement is used to decease the random error of single subaperture. The subaperture positioning is also carefully corrected to ensure the pixels of the adjacent subapertures in overlapping areas can be matched well. A first stitching measurement result of a spherical mirror with 30 meters radius is shown.
Conference Presentations are recordings of oral presentations given at SPIE conferences and published as part of the conference proceedings. They include the speaker's narration along with a video recording of the presentation slides and animations. Many conference presentations also include full-text papers. Search and browse our growing collection of more than 14,000 conference presentations, including many plenary and keynote presentations.
Study of self-shadowing effect as a simple means to realize nanostructured thin films and layers with special attentions to birefringent obliquely deposited thin films and photo-luminescent porous silicon