The 488nm laser is the most important excitation light source of flow cytometry. The indirect diode (frequency-doubled diode) 488nm lasers are used in the excitation of Becton Dickinson (BD) AccuriTM C6. For using cost effective lasers, we have validated direct diode 488nm lasers as the replacement component of frequency-doubled diode laser. BD Bioscience issued the protocols to cover wavelength, power, noise, and polarization at the operation temperature range of cytometer. Pavilion Integration Corporation (PIC) tested 6 samples as the component validation of direct diode 488nm lasers based on the protocols from BD Biosciences. BD Bioscience also tested one of laser samples to further validate the test results of power, noise, and polarization from PIC.
Wei P. Chen and Ningyi D. Luo, "The component validation of direct diode 488nm lasers in BD Accuri C6 flow cytometers," Proc. SPIE 9700, Design and Quality for Biomedical Technologies IX, 97000E (Presented at SPIE BiOS: February 13, 2016; Published: 18 March 2016); https://doi.org/10.1117/12.2213432.
Conference Presentations are recordings of oral presentations given at SPIE conferences and published as part of the proceedings. They include the speaker's narration with video of the slides and animations. Most include full-text papers. Interactive, searchable transcripts and closed captioning are now available for 2018 presentations, with transcripts for prior recordings added daily.
Search our growing collection of more than 16,000 conference presentations, including many plenaries and keynotes.