ISM can be performed in fluorescence, bright field or interference microscopy. Several different implementations have been described, with associated advantages and disadvantages. In two-photon microscopy, the illumination and detection point spread functions are very different. This is also the case when using pupil filters or when there is a large Stokes shift.
ACCESS THE FULL ARTICLE
SEARCH WITHIN PRESENTATIONS
CONFERENCE NAME -
CONTENT TYPE -
Please refine your search query.