From Event: SPIE Optical Engineering + Applications, 2016
In the near-infrared-ray computed tomography (NIR-CT) scanner, NIR rays are produced from a light-emitting diode
(LED) and detected using an NIR phototransistor (PT). The wavelengths of the LED peak intensity and the PT high
sensitivity in the data table are both 940 nm. The photocurrents flowing through the PTR are converted into voltages
using an emitter-follower circuit, and the output voltages are sent to a personal computer through an analog-digital
converter. The NIR projection curves for tomography are obtained by repeated linear scans and rotations of the object,
and the scanning is conducted in both directions of its movement.
Conference Presentations are recordings of oral presentations given at SPIE conferences and published as part of the conference proceedings. They include the speaker's narration along with a video recording of the presentation slides and animations. Many conference presentations also include full-text papers. Search and browse our growing collection of more than 12,000 conference presentations, including many plenary and keynote presentations.