From Event: SPIE Optical Engineering + Applications, 2016
This paper presents best practices for modeling the resolution of dispersive imaging spectrometers. The differences between sampling, width, and resolution are discussed. It is proposed that the spectral imaging community adopt a standard definition for resolution as the full-width at half maximum (FHWM) of the total line spread function (LSF). Resolution should be computed for each of the spectral, cross-scan spatial, and along-scan spatial/temporal dimensions separately. A physical optics resolution model is presented that incorporates the effects of slit diffraction and partial coherence, the result of which is a narrower slit image width and reduced throughput.
John F. Silny, "Resolution modeling of dispersive imaging spectrometers," Proc. SPIE 9976, Imaging Spectrometry XXI, 99760A (Presented at SPIE Optical Engineering + Applications: August 29, 2016; Published: 19 September 2016); https://doi.org/10.1117/12.2239872.
Conference Presentations are recordings of oral presentations given at SPIE conferences and published as part of the conference proceedings. They include the speaker's narration along with a video recording of the presentation slides and animations. Many conference presentations also include full-text papers. Search and browse our growing collection of more than 14,000 conference presentations, including many plenary and keynote presentations.