An optical profilometer that uses a Techmet LaserMike scanning, focused, laser-beam, optical micrometer is installed in a remote alpha-gamma containment cell at the Los Alamos Hot-Cell Facility.1 A hot-cell extension chamber provides the nominal 30-cm (12-in.) working distance required by the LaserMike and, at the same time, keeps the LaserMike components outside the high-radiation-containment environment. This system provides measurement accu-racy better than±5 pm (0.0002 in.) on diameters between 2 and 13 mm (0.08 and 0.5 in.) at a rate of 33 measurements per second. The Hot-Cell Facility also uses a Korad 20-J-output ruby pulsed laser to drill a hole in reactor fuel element cladding to sample fission gas. The laser is then used to reweld the hole so that the fuel element will not be contaminated and may be stored without an alpha-containment barrier. The wall thickness of the fuel elements sampled varies from 0.25 to 0.50 mm (0.010 to 0.020 in.).