The absolute measurement technique for atmospheric temperature can avoid the calibration process and improve the measurement accuracy. To achieve the rotational Raman temperature lidar of absolute measurement, the two-stage parallel multi-channel spectroscopic filter combined a first-order blazed grating with a fiber Bragg grating is designed and its performance is tested. The parameters and the optical path structure of the core cascaded-device (micron-level fiber array) are optimized, the optical path of the primary spectroscope is simulated and the maximum centrifugal distortion of the rotational Raman spectrum is approximately 0.0031 nm, the centrifugal ratio of 0.69%. The experimental results show that the channel coefficients of the primary spectroscope are 0.67, 0.91, 0.67, 0.75, 0.82, 0.63, 0.87, 0.97, 0.89, 0.87 and 1 by using the twelfth channel as a reference and the average FWHM is about 0.44 nm. The maximum deviation between the experimental wavelength and the theoretical value is approximately 0.0398 nm, with the deviation degree of 8.86%. The effective suppression to elastic scattering signal are 30.6, 35.2, 37.1, 38.4, 36.8, 38.2, 41.0, 44.3, 44.0, 46.7 dB. That means, combined with the second spectroscope, the suppression at least is up to 65 dB. Therefore we can fine extract single rotational Raman line to achieve the absolute measurement technique.