Since the modeling of machining processes relies on high-strain-rate, high-temperature material properties, NIST has built a split-Hopkinson (or Kolsky) bar to determine the stress-strain behavior of rapidly heated materials at high temperatures. Our Kolsky bar has been constructed in the NIST high current pulse-heating facility, which enables electrically heating the samples within ~ 100 milliseconds time duration, immediately before the mechanical impact in the bar. Due to the rapid heating, we avoid possible structural changes in the sample, and a stress-strain relationship can be determined at different temperatures for various test materials. We describe the design and the development of the resistively-heated Kolsky-bar apparatus. The incident and the transmitted bars are constructed of 1.5 m long, 15 mm diameter maraging steel, and a typical sample is a 4 mm-diameter, 2 mm-long cylinder of 1045 steel. The sample is placed between the bars and held by friction. The current is transmitted through the graphite-sleeve bushings of the two bars. The non-contact temperatures are measured using an InGaAs near-infrared micro-pyrometer (NIMPY) and an InSb focal-plane (320 by 256) array (thermal camera). The NIMPY and the thermal camera are both calibrated using a variable-temperature blackbody, and the thermodynamic temperature of the metal is determined using the emissivity determined from the measured infrared spectral reflectance of the metal. Thermal videos of the electrically-heated and the room-temperature impacts will be shown with 1 kHz frame rates, and the changes in the stress-strain curves with the temperature of the samples will be discussed.