In the test course of static transfer function of TDI CCD camera, because of the influence that gets environmental and artificial etc. factor, the value of static transfer function
measured at any time is between unceasing fluctuation, so, make accuracy reduce. To solve
this problem, a kind of accurate measurement technique of static transfer function is put forward. First, before carrying out the measure of static quiet of transfer function, the best test point of transfer function of the TDI CCD camera must be determined, it is parallel to guarantee the rectangle target surface of parallel optical pipe and camera focal plane
maintenance parallel, and again guarantee target strip in rectangle target and TDI CCD in camera focal plane maintenance vertical. TDI CCD catches rectangle target image, per 1000
lines of target mark image as a measures sample of static transfer function, exclude because of atmosphere tremble twisted, vague rectangle target mark image, retain 500 distinct and steady
target mark image as measure sample set. Then, calculate the static transfer function of each measure sample respectively, take the average of all static quiet transfer function in measure
sample set as the static transfer function of camera. Finally, the measure of the static transfer function for TDI CCD camera makes error analysis. Experimental results indicate that the value of the static transfer function of TDI CCD camera measured with this kind of method is 0.2923, with before measurement technique comparison, the value of static transfer function has raised 0.02, makes the accuracy of the measure of static transfer function have gotten raising.