Pyranometer is an instrument used to measure the solar radiation, according to pyranometer differs as installation state, can be respectively measured total solar radiation, reflected radiation, or with the help of shading device for measuring scattering radiation. Pyranometer uses the principle of thermoelectric effect, inductive element adopts winding plating type multi junction thermopile, its surface is coated with black coating with high absorption rate. Hot junction in the induction surface, while the cold junction is located in the body, the cold and hot junction produce thermoelectric potential. In the linear range, the output signal is proportional to the solar irradiance. Traceability to national meteorological station, as the unit of the national legal metrology organizations, the responsibility is to transfer value of the sun and the earth radiation value about the national meteorological industry. Using the method of comparison, with indoor calibration of solar simulator, at the same location, standard pyranometer and measured pyranometer were alternately measured radiation irradiance, depending on the irradiation sensitivity standard pyranometer were calculated the radiation sensitivity of measured pyranometer. This paper is mainly about the design and calibration method of the pyranometer indoor device. The uncertainty of the calibration result is also evaluated.