Access to eBooks is limited to institutions that have purchased
or currently subscribe to the SPIE eBooks program. eBooks are not
available via an individual subscription. SPIE books (print and
digital) may be purchased individually on
Contact your librarian to recommend SPIE eBooks for your organization.
Two other topics that should not be overlooked are laser safety and caring for the optics in your microscope. Here is a summary of cautions to take when using lasers and light microscopes.
15.1 Laser Safety
â¢ Read one or more source books on laser safety.
â¢ Be aware of the different types of lasers, their damage thresholds, and the risks of ocular damage, skin damage, and fire associated with each type of laser.
â¢ Be aware of the advantages (ocular protection) and disadvantages (no knowledge of position of the laser beam) of laser safety goggles.
â¢ When designing a laser system on an optical table be sure that the laser beam is not at eye height.
â¢ Be aware of the fire danger with lasers. Most materials in the clinic are flammable.
15.2 How to Clean Optics
â¢ Optics should be cleaned at the following times: when there is a power loss; at regular maintenance intervals; when a laser cavity is opened; when a new set of mirrors is installed; when new optics are received.
â¢ Learn how to correctly clean a microscope objective prior to and after use.
â¢ Particles of dust can severely degrade the image quality of optical components.
â¢ Improper cleaning of a microscope objective will cause microscopic scratches that will permanently degrade the image quality. Never use anything but lens-cleaning paper with spectroscopic-grade acetone and methanol. Note that these solvents can solubilize oil from your skin and transfer it to the surface of the optical element.
â¢ Use a dust cover made from a conductive material to shield the microscope from dust.
Online access to SPIE eBooks is limited to subscribing institutions.