Stokes V, the circular polarization of light, from the solar corona is weak of the order of 10 − 4 times the intensity (Stokes I). Measuring weak source polarization, for a faint source such as corona, is difficult and requires long integration time. To obtain long uninterrupted measurements, a space-based polarimeter would be preferable over a ground-based observatory. A full-Stokes polarimeter is designed such that the modulation matrix provides high efficiency in measuring the weak Stokes V signal, while minimizing the cross talk from Stokes Q and Stokes U. The prototype polarimeter consists of a single crystal retarder as the modulator and a Wollaston prism as a dual beam analyzer. The modulation is performed by rotating the retarder continuously. An optimum modulation matrix is derived taking into account the systematics and polarization cross talk due to satellite jitter. We present the results of cross talk simulation and the steps taken to obtain the modulation matrix. A polarimeter designed to observe the solar corona at Fe XIII 1074.6-nm emission line is presented. Jitter and drift from a low earth orbit satellite are taken to simulate and experimentally verify the cross talk and polarimetric efficiency of the polarimeter. A plane parallel crystal retarder produces polarized fringes. F-ratio of the beam incident on the retarder is one of the factors that effects the contrast of the polarized fringes. Results from a simulation performed to compare the polarizance produced by the retarder, when placed in a converging beam and a collimated beam of light, is presented.
You have requested a machine translation of selected content from our databases. This functionality is provided solely for your convenience and is in no way intended to replace human translation. Neither SPIE nor the owners and publishers of the content make, and they explicitly disclaim, any express or implied representations or warranties of any kind, including, without limitation, representations and warranties as to the functionality of the translation feature or the accuracy or completeness of the translations.
Translations are not retained in our system. Your use of this feature and the translations is subject to all use restrictions contained in the Terms and Conditions of Use of the SPIE website.