In vivo Cerenkov luminescence imaging (CLI) is a rapidly growing molecular imaging research field based on the detection of Cerenkov radiation induced by beta particles when traveling though biological tissue with a velocity greater then the speed of light in the tissue. In the past two years several papers have been published investigating the use of Cerenkov radiation to image different radiotracers,188.8.131.52.6.–7 as an excitation source for quantum dots,89.–10 and as a light source for phototherapy.11
Generally speaking, all these studies pay little or no attention to exploiting the peculiar properties of the Cerenkov light spectrum. For example, in our previous work2,4 the shape of the Cerenkov spectrum has been used to obtain, respectively, the source depth or to reconstruct 3-D images using a multispectral approach.14
It is well known that the number of the emitted Cerenkov photons per unit of wavelength shows a dependence,15 and thus a larger number of photons are emitted at shorter wavelengths and vice versa. This property was used by Ref. 16 to investigate theoretically the possible use of fiber probes for brain mapping in order to avoid the typical absorption of the short wavelength part of the spectrum in the biological tissues. Their theoretical analysis showed that the use of fiber probes can significantly increase by three orders of magnitude the detected Cerenkov signal.
However there are some drawbacks of using an approach based on fiber probes, such as its invasiveness and the fact that it is also not possible to obtain continuous images of the Cerenkov sources as in CLI or CLT.
In this work we propose a quite opposite and, to some extent, counterintuitive approach to enhance the number of the detected Cerenkov photons. More precisely, our main goal was to investigate the detection of Cerenkov photons in the near infrared (NIR) region of the spectrum where the tissues show a smaller absorption coefficient.
In the rest of this short communication we will present a theoretical analysis based on applying a photon propagation diffusion model to Cerenkov photons in the tissue. The detection of the photons exiting the tissue will then be simulated considering charge-coupled device (CCD) detectors optimized, respectively, for the visible and NIR spectral regions.
We will show that despite the smaller number of Cerenkov photons in the NIR, the fraction exiting the tissues is greater than in the visible range, and thus, the use of a CCD detector optimized for the NIR range will allow for improvement of the signal.
The photons propagation model used in this work was developed by Haskel and colleagues17 and applied to preclinical small animal optical imaging by Rice and collaborators.18 The photon transport in a turbid medium with an absorption coefficient can be described by the diffusion equation
Since we are interested in a noninvasive imaging approach, we assumed that the detector is located outside the tissue and, thus, the whole system geometry can be approximated as a planar semi-infinite medium.
This assumption is more valid for small animal optical systems where the animals are squeezed by two slabs of transparent material.
The semi-infinite medium boundary condition can be included in solving the diffusion equation and for a point source in the case of a detector placed above the medium, the surface radiance at the tissue surface is equal to
At this point it may be useful to remember that the emission of Cerenkov radiation with a radial frequency takes place when the ratio between the particle velocity and the speed of light is15,19 one can obtain the total energy emitted by an electron with charge travelling a short length in a medium as
The corresponding emitted power is thus equal to10) can be combined with Eq. (3) in order to estimate the power of Cerenkov radiation emitted by a beta particle traveling in the tissues.
The model described above relies on the assumption that the velocity of the beta particle is almost constant along and, at the same time, such distance needs to be small since Eq. (3) strictly speaking is valid for a point source.
It is useful to remind here that CLI imaging is based on beta emitters used in nuclear medicine these radioisotopes have an endpoint energy typically below 2 MeV, and thus the particles travel a relatively short distance in the medium. More precisely, a recent paper20 based on Monte Carlo simulation showed that the Cerenkov light production is confined to 2 mm (root mean squared) even for the isotopes with high endpoint energies. For the widely used 18F the Cerenkov light production take place only within 0.3 mm of the positron path.
In Ref. 18 are presented the full width half maximum (FWHM) values for a point source located at different depths; these FWHM values are generally greater with respect to the distance where Cerenkov light production take places. Therefore for our purposes we can consider the Cerenkov source to be almost equivalent to a point-like source.
The measured radiance will be dependent on the quantum efficiency of the detector, more precisely21,22
In order to investigate the differences in the measured radiance, the profiles of two different commercial CCD were included in Eq. (11).
In order to compare the detected signals, a figure of merit was calculated as23,24
As an example, the spectrums of the detected Cerenkov light emitted from a point source located at 3-mm depth is shown in Fig. 1. The three plots were obtained by considering an ideal detector having across the entire spectrum, and the two values presented in Refs. 23, 24. As one can see by looking at Fig. 1, the spectrum obtained using a CCD more efficient in the NIR region is closer to the spectrum obtained considering an ideal detector.
was calculated at different source depths ranging from 1 to 11 mm and plotted in Fig. 2. The plot present a monotonic increase of with respect to source depth, showing that a CCD optimized for the NIR always allows a better detection of the Cerenkov radiation exiting the animal tissues.
Considering only the Poisson nature of the noise, the relative differences in the signal to noise ratio (SNR) are . For example, a value of corresponds to a 16% improvement of the SNR.
Of course there are other important sources of noise like the CCD reading noise and dark current. These sources of noise are dependent on the detector temperature, electronics, and reading mode. A detailed analysis of all these noise sources is beyond the main goals of this short note.
The theoretical analysis presented here shows that the use of a CCD optimized for the NIR allows a better detection of the Cerenkov photons exiting the animal surface, suggesting that small animal optical imaging systems tailored on the NIR range perform well for CLI.
We would like to stress that our analysis is based on several assumptions and approximations like considering almost constant the particle speed and Cerenkov radiation emitted by a point source. However, we believe that these are not severe limiting factors. More precisely, our main goal was to compare in relative terms [see Eq. (12)] the efficiency of Cerenkov photons detection in the visible and NIR region and not to obtain absolute signal quantitation. The comparison was performed considering Cerenkov point sources located at different depths inside the animal.
We conclude that noninvasive CLI imaging can be improved by using small animal optical imaging systems based on a CCD detector with a quantum efficiency peaked in the NIR range. The improvement is more significant when the Cerenkov source to be imaged is located deeper inside the animal.
R. Robertsonet al., “Optical imaging of Cerenkov light generation from positron-emitting radiotracers,” Phys. Med. Biol. 54(16), N355–N365(2009).PHMBA70031-9155http://dx.doi.org/10.1088/0031-9155/54/16/N01Google Scholar
A. E. Spinelliet al., “Cerenkov radiation allows in vivo optical imaging of positron emitting radiotracers,” Phys. Med. Biol. 55(2), 483–495 (2010).PHMBA70031-9155http://dx.doi.org/10.1088/0031-9155/55/2/010Google Scholar
A. E. Spinelliet al., “Cerenkov radiation imaging of beta emitters: in vitro and in vivo results,” Nucl. Instrum. Meth. A 648(Suppl. 1), S310–S312 (2011).NIMAER0168-9002http://dx.doi.org/10.1016/j.nima.2010.11.038Google Scholar
F. Boschiet al., “In vivo (18)F-FDG tumour uptake measurements in small animals using Cerenkov radiation,” Eur. J. Nucl. Med. 38(1), 120–127 (2011).EJNMD90340-699710.1007/s00259-010-1630-yGoogle Scholar
A. E. SpinelliF. Boschi, “Unsupervised analysis of small animal dynamic Cerenkov luminescence imaging,” J. Biomed. Opt. 16(12), 12 (2011).JBOPFO1083-3668http://dx.doi.org/10.1117/1.3663442Google Scholar
R. S. Dothageret al., “Cerenkov radiation energy transfer (CRET) imaging: a novel method for optical imaging of PET isotopes in biological systems,” PLoS One 5(10), e13300 (2010).1932-620310.1371/journal.pone.0013300Google Scholar
C. Ranet al., “In vivo photoactivation without “light”: use of Cherenkov radiation to overcome the penetration limit of light,” Mol. Imag. Biol. 14(2), 156–162 (2012).1860200210.1007/s11307-011-0489-zGoogle Scholar
Z. Huet al., “Experimental Cerenkov luminescence tomography of the mouse model with SPECT imaging validation,” Opt. Express 18(24), 24441–24450 (2010).OPEXFF1094-4087http://dx.doi.org/10.1364/OE.18.024441Google Scholar
A. E. Spinelliet al., “Multispectral Cerenkov luminescence tomography for small animal optical imaging,” Opt. Express 19(13), 12605–12618 (2011).OPEXFF1094-4087http://dx.doi.org/10.1364/OE.19.012605Google Scholar
J. V. Jelley, Cerenkov Radiation and Its Applications, Pergamon, London (1958).Google Scholar
A. ZheltikovK. Anokhin, “Fiber-probe detection for positron-emission-assisted Cherenkov-radiation brain mapping,” Phys. Rev. E 6, 061902–061907 (2011).PLEEE81063-651Xhttp://dx.doi.org/10.1103/PhysRevE.84.061902Google Scholar
R. C. Haskellet al., ‘‘Boundary conditions for the diffusion equation in radiative transfer,’’ J. Opt. Soc. Am. A 11(10), 2727–2741 (1994).JOAOD60740-3232http://dx.doi.org/10.1364/JOSAA.11.002727Google Scholar
P. A. Cerenkov, Radiation of Particles Moving at a Velocity Exceeding That of Light, and Some of the Possibilities for Their Use in Experimental Physics, Nobel Lectures, Physics, Elsevier Publishing, Amsterdam, 1942–1962 (1964).Google Scholar
G. S. Mitchellet al., “In vivo Cerenkov luminescence imaging: a new tool for molecular imaging,” Phil. Trans. A Math. Phys. Eng. Sci. 369(1955), 4605–4619 (2011).1364-503Xhttp://dx.doi.org/10.1098/rsta.2011.0271Google Scholar
C. Kuoet al., “Three-dimensional reconstruction of in vivo bioluminescent sources based on multispectral imaging,” J. Biomed. Opt. 12(2), 024007 (2007).JBOPFO1083-3668http://dx.doi.org/10.1117/1.2717898Google Scholar
Living Image 4.0, Caliper Life Sciences, http://www.caliperls.com/products/precl.inical-imaging/living-image.htm (last accessed 20 Feb 2011).Google Scholar
http://www.caliperls.com/products/preclinical-imaging/ivis-imaging-system-200-series.htm (last accessed 20 Feb 2011).Google Scholar
http://www.andor.com/pdfs/literature/Andor_iKon-M_Flyer.pdf (last accessed 20 Feb 2011).Google Scholar