## 1.

## Introduction

During the past century, the attraction of biophysicists for observing and characterizing living matter at cellular and subcellular levels has been the prime mover for developing sophisticated microscopic devices. Even if the concept of diffraction-limited imaging and interferometry principles^{1} was established for a long time, the actual development of microscopic devices based on interferometric contrast was achieved in the first part of the 20th century by Zernike^{2} and Nomarski and Weill^{3} and further developed by Gabor,^{4} who pioneered the principle of holographic microscopy. Retrieving a phase information from the light transmitted through transparent objects, like living cells, has benefited from the development of coherent sources, optoelectronic polarizing tools, and fast and sensible cameras in the second part of the 20th century. It is nowadays quite straightforward to design a compact, highly sensitive phase microscope that can follow in real time the dynamics of cells. In the last decade, different teams^{5}6.7.8.9.10.11.12.^{–}^{13} have played a major role in disseminating the concepts of quantitative phase microscopy (QPM) among the optical and biophysical community. They have applied this technique to the real-time characterization of cellular dynamics and their alteration in cases of diseases.^{6}^{,}^{13}14.15.^{–}^{16}

Other approaches that do not rely on interferometric principles have also been proposed to circumvent the constraint of high degree of coherence. They rather start from a principle of electromagnetic energy conservation, written as the divergence of the wave flow vector being equal to zero.^{17}18.19.20.^{–}^{21} The technique of transport of intensity is valid only for weak defocus and reduces to a differential equation for field propagation.^{22}^{,}^{23} This method is interesting for partially coherent illumination such as given by broadband polychromatic sources. For thick samples, three-dimensional computation is required.^{24}

The principle of diffraction phase microscopy (DPM) introduced by Popescu and coauthors^{25}26.^{–}^{27} relies on both off-axis and common-path principles in combination with fast acquisition rate and high temporal sensitivity. The interference patterns produced by a DPM system correspond to the superimposition of a simple carrier fringe pattern, given, for instance, by a diffraction grating, with the image of the object through the objective lens. To retrieve the phase image associated with the sample object, different methods have been proposed, including Hilbert transform followed by phase unwrapping,^{28}^{,}^{29} derivative methods,^{30} and Fourier filtering to avoid unwrapping problems.^{31} All these phase retrieval algorithms rely on the assumption that the object phase does not alter the fringe carrier pattern, allowing a quasi one-dimensional analysis of interference patterns. To improve this approach and delineate more precisely the validity of this assumption, we propose here to generalize Fourier filtering methods using a two-dimensional (2-D) space-scale analysis based on Morlet wavelet transform.^{32} We first introduce the DPM principle and illustrate it on simple physical objects, such as micron-size particles. We then describe the space-scale analysis and the possibility to measure directly the phase of the interference pattern from the detection of the ridges of the Morlet wavelet transform of the original image. Finally, we illustrate this method on living myoblasts, showing that both the modulus and the phase of the interference pattern can be retrieved.

## 2.

## Modeling the QPM System Response and Its Space-Scale Analysis

Popescu and coauthors^{25}26.^{–}^{27} designed an elegant QPM by introducing an amplitude diffraction grating in the image plane of an inverted microscope to generate multiple diffraction orders containing the full spatial information about the transparent object crossed by the light beam. This QPM setup, inspired from this principle, was assembled in our laboratory. It is described in the Sec. 7 and illustrated in Fig. 1. After the transmission grating G, the zeroth- (${U}_{0}$) and first-order (${U}_{1}$) components of the diffracted beams are separated in the conjugated Fourier plane of the image plane of the microscope. Then, the zeroth order is low-pass filtered with a spatial filter and recombined with the first order, thanks to a second Fourier lens ${L}_{2}$ to give a spatially modulated interference image $I(x,y)$ similar to the one shown in Fig. 2(a), captured from a glass coverslip covered by a scratched polymer layer. The vertical fringes of this image have a frequency ${f}_{g}$ that is related to the grating period; they superimpose to the object phase image. We propose here a method based on a 2-D continuous wavelet transform (CWT) to retrieve the object phase information from this type of image.

The intensity map $I(x,y)$ recorded on the CMOS camera is directly proportional to the modulus square of the electric field at this point.

## (1)

$$I(\overrightarrow{x})={|{U}_{0}+{U}_{1}|}^{2}(\overrightarrow{x})=[{|{U}_{0}|}^{2}+{|{U}_{1}|}^{2}+{U}_{0}{U}_{1}^{*}+{U}_{0}^{*}{U}_{1}](\overrightarrow{x}).$$If ${f}_{g}$ is the spatial frequency of the grating, the phase difference between ${U}_{1}$ and ${U}_{0}$ includes both the grating and the object phase information: $\mathrm{\Phi}(\overrightarrow{x})={f}_{g}x+\varphi (\overrightarrow{x})$, with $\overrightarrow{x}=(x,y)$. This gives a synthetic form of $I(\overrightarrow{x})$.

## (2)

$$I(\overrightarrow{x})=P(\overrightarrow{x})+Q(\overrightarrow{x})\mathrm{cos}[{f}_{g}x+\varphi (\overrightarrow{x})].$$$\varphi (\overrightarrow{x})$ is the phase due to the object transmission at location $\overrightarrow{x}$. $P(\overrightarrow{x})$ and $Q(\overrightarrow{x})$ are real valued; they correspond, respectively, to the background and modulation intensities at location $\overrightarrow{x}$. One common assumption is that $P(\overrightarrow{x})$ and $Q(\overrightarrow{x})$ vary much slower than $\varphi (\overrightarrow{x})$. The local frequencies of the signal $U$ can be computed in both directions $x$ and $y$ from the partial derivatives of the phase $\varphi (\overrightarrow{x})$.

These equations can be rewritten in the vectorial form.

## (5)

$$\left[\begin{array}{c}{f}_{x}\\ {f}_{y}\end{array}\right]=\overrightarrow{\nabla}[{f}_{g}x+\varphi (\overrightarrow{x})].$$Equation (3) shows that the local frequency in $x$ may deviate from the carrier frequency ${f}_{g}$ depending on the strength of the phase derivative with respect to $x$, the steeper $\varphi (\overrightarrow{x})$, the larger this deviation.

By a simple computation, assuming that we are characterizing a homogeneous object of thickness $d$ and index $n$ imbedded in a continuum medium of index ${n}_{0}$, we can approximate the optical path change $\mathrm{\Delta}{p}_{o}$ at the center of the object with the relation $\mathrm{\Delta}{p}_{o}\sim d(n-{n}_{0})$ and $\mathrm{\Delta}\varphi =2\pi \mathrm{\Delta}{p}_{o}/\lambda $. Assuming that the microscope can achieve a ${10}^{-2}$ rad phase sensitivity, this gives the possibility to detect objects of thickness ${10}^{-2}\lambda /(2\pi \mathrm{\Delta}n)$, which for a refractive index drop of 0.2 and wavelength $\lambda =532\text{\hspace{0.17em}}\text{\hspace{0.17em}}\mathrm{nm}$ gives a sensitivity to objects with thickness down to 4 nm, along the optical axis.

The practical treatment of interference fringe patterns often assumes that the background and the fringe modulation intensities as well as the phase vary slowly across the fringe pattern.^{33}^{,}^{34} This may no longer be true when the fringe pattern is produced by a highly diffracting object, such as a highly structured living cell, for instance, with thickness of several micrometers. A living cell is not a homogeneous medium, but is made of compartments surrounded by lipid membranes with high refractive index (plasmic membrane, nuclear membrane, Golgi apparatus, mitochondrial network) and highly dynamic proteic fiber network (cytoskeleton, nuclear matrix). This situation requires a method that is able to capture the spatial variation of the local frequencies ${f}_{x}$ and ${f}_{y}$, without being biased by the spatial dependence of the fringe amplitude $Q(\overrightarrow{x})$ [Eq. (2)]. The wavelet transform offers this possibility since it allows a decomposition of a signal (uni- or multidimensional) into atoms (wavelets) that are well localized in space and frequency.^{35} The CWT is a mathematical technique introduced in signal analysis in the early 1980s.^{36}^{,}^{37} Since then, it has been the subject of considerable theoretical developments and practical applications in a wide variety of fields.^{38}39.40.41.42.43.44.45.46.47.48.49.50.51.52.53.^{–}^{54} An optical wavelet device has also been designed that performs the CWT, thanks to Fourier optics principles.^{55}^{,}^{56} We choose here a 2-D Morlet wavelet^{32} because it is particularly well suited for time (or space) frequency analysis.

## 3.

## Phase Map Reconstruction Using the CWT

Phase stepping and Fourier transform methods have been proposed in the 1980s for interferogram analysis in one and two dimensions.^{33}^{,}^{57} These methods, however, assume that the coefficients $P(\overrightarrow{x})$ and $Q(\overrightarrow{x})$ vary slowly over one fringe cycle, so they can be treated as constants. To our knowledge, the first attempt to use a wavelet transform as a processing tool with white-light interferometry dates back to 1997.^{58} Since then many teams have used this tool with success, in one dimension^{59}60.61.62.63.64.65.66.^{–}^{67} as well as in two dimensions.^{68}69.70.71.72.73.74.^{–}^{75}

More than a simple signal filtering tool, the wavelet transform analysis can also offer the possibility of a direct computation of the phase of the interferometry image, thanks to a wavelet transform ridge detection algorithm.^{59}60.61.^{–}^{62}^{,}^{64}^{,}^{66}^{,}^{67}^{,}^{75}

The 2-D CWT of an interferogram $I(\overrightarrow{x})$ with $\overrightarrow{x}=(x,y)$ is defined as^{52}

## (6)

$${W}_{\mathrm{\Psi}}(\overrightarrow{b},a,\theta )={a}^{\eta}{\int}_{{\mathfrak{R}}^{2}}I(\overrightarrow{x}){\mathrm{\Psi}}^{*}[{a}^{-1}{r}_{\theta}(\overrightarrow{x}-\overrightarrow{b})]{d}^{2}\overrightarrow{x}.$$${W}_{\mathrm{\Psi}}(\overrightarrow{b},a,\theta )$ is the wavelet transform coefficient, at position $\overrightarrow{b}$, scale parameter $a$, and rotation angle $\theta $. $\overrightarrow{b}=({b}_{x},{b}_{y})$ is a 2-D translation parameter describing the position of the wavelet, $a>0$ is the scale dilation parameter (nondimensioned), $\theta $ is a rotation parameter, ${\mathbf{r}}_{\theta}$ is the $2\times 2$ rotation operator matrix, $\mathrm{\Psi}$ is the mother wavelet, ${\mathrm{\Psi}}^{*}$ is the complex conjugate of $\mathrm{\Psi}$, and $\eta $ is a normalization exponent. In Fourier space, the wavelet transform reads as

## (7)

$${W}_{\mathrm{\Psi}}(\overrightarrow{b},a,\theta )={a}^{\eta}{\int}_{{\mathfrak{R}}^{2}}\hat{I}(\overrightarrow{k}){e}^{i\overrightarrow{b}\xb7\overrightarrow{k}}{\hat{\mathrm{\Psi}}}^{*}[{ar}_{-\theta}(\overrightarrow{k})]{d}^{2}\overrightarrow{x}.$$The symbol $\widehat{}$ denotes the Fourier transformation. A typical mother wavelet commonly used to detect localized and oriented features is the 2-D Morlet wavelet.^{52}

## (8)

$${\mathrm{\Psi}}_{M}(\overrightarrow{x})={e}^{i{\overrightarrow{k}}_{0}\xb7\overrightarrow{x}}{e}^{-\frac{1}{2}{|A\overrightarrow{x}|}^{2}}-{e}^{-\frac{1}{2}{|{A}^{-1}{\overrightarrow{k}}_{0}|}^{2}-\frac{1}{2}{|A\overrightarrow{x}|}^{2}},$$## (9)

$${\widehat{\mathrm{\Psi}}}_{M}(\overrightarrow{k})=\sqrt{\epsilon}[{e}^{-\frac{1}{2}{|{A}^{-1}(\overrightarrow{k}-{\overrightarrow{k}}_{0})|}^{2}}-{e}^{-\frac{1}{2}{|{A}^{-1}{\overrightarrow{k}}_{0}|}^{2}-\frac{1}{2}{|{A}^{-1}\overrightarrow{k}|}^{2}}].$$The parameter ${k}_{0}$ is the wave vector and $A=diag[1,{\epsilon}^{1/2}]$ is a $2\times 2$ anisotropic matrix ($\epsilon \ge 1$).

We use here this anisotropic 2-D continuous Morlet wavelet transform, with anisotropy factor $\u03f5$, to extract the phase of a fringe pattern obtained with the QPM, such as shown in Fig. 2(a). The correction terms in Eqs. (8) and (9) enforce the admissibility condition ${\widehat{\mathrm{\Psi}}}_{M}(\overrightarrow{0})=0$. However, they are numerically negligible for $|{k}_{0}|\ge 5.6$, and one usually drops them. Putting $\epsilon =1$ and removing the correction terms gives the Gabor function.

## (10)

$${\mathrm{\Psi}}_{G}(\overrightarrow{x})=\mathrm{exp}(i{\overrightarrow{k}}_{0}\xb7\overrightarrow{x})\mathrm{exp}(-\frac{1}{2}{|\overrightarrow{x}|}^{2}).$$The Morlet wavelet is a complex function; the modulus of the truncated Morlet wavelet (without the correction term) is a Gaussian, elongated in the $x$ direction if $\epsilon >1$, and its phase is constant along the direction orthogonal to ${\overrightarrow{k}}_{0}$, and linear in $\overrightarrow{x}$, mod($2\pi /|{k}_{0}|$), along the direction of ${\overrightarrow{k}}_{0}$. As compared to the 1-D case, the additional feature of the 2-D Morlet (or Gabor) wavelet function is its inherent directivity, entirely contained in its phase. This turns to be a crucial advantage for studying objects with directional properties. Indeed, since the wavelet transform [Eq. (6)] is a convolution product of the fringe pattern with the dilated wavelet, we see that the wavelet transform smoothes the image in all directions, but detects the sharp transitions in the direction perpendicular to ${\overrightarrow{k}}_{0}$. In Fourier space, the effective support (footprint) of the function ${\widehat{\mathrm{\Psi}}}_{M}$ is an ellipse centered at ${\overrightarrow{k}}_{0}$ and elongated in the ${k}_{y}$ direction. In Figs. 2(b) and 2(c) we show two Morlet wavelets computed for ${\overrightarrow{k}}_{0}=(5.6,0)$, and $\epsilon =1$ and $\epsilon =10$, respectively; their Fourier transforms are shown in Figs. 2(e) and 2(f). Since the ratio of the axes is equal to $\sqrt{\epsilon}$, the cone of the wavelet in Fourier space elongates along ${k}_{y}$ direction as $\epsilon $ increases. This wavelet preferentially detects edges perpendicular to the $y$-direction (i.e., parallel to ${\overrightarrow{k}}_{0}$), and its angular selectivity increases with ${\overrightarrow{k}}_{0}$ and with the anisotropy $\epsilon $. For the optical image shown in Fig. 2(a) recorded with the QPM from a glass coverslip coated with a scratched polymer layer, the best selectivity is achieved with ${\overrightarrow{k}}_{0}$ perpendicular to the long axis of the ellipse in $\overrightarrow{k}-space$, that is ${\overrightarrow{k}}_{0}=({k}_{0},0)$. We show in Fig. 2(d) the modulus of the Fourier transform of the fringe image shown in Fig. 2(a). The Morlet wavelet selects the right part of this Fourier transform by performing a band-pass filtering around the grating frequency. The advantage of taking a smooth wavelet and not a simply circular window in Fourier space^{31} is not only to avoid the introduction of artificial oscillations produced by the sharp boundary of such a window, but also to have the ability to use the mathematical formalism of wavelet analysis, for instance, the ridge detection method.^{52} The Morlet wavelet ${\mathrm{\Psi}}_{M}$ is then written as

## (11)

$${\mathrm{\Psi}}_{M}(\overrightarrow{x})=\mathrm{exp}[-\frac{1}{2}({x}^{2}+\epsilon {y}^{2})][\mathrm{exp}(i{k}_{0}x)-\mathrm{exp}(-{k}_{0}^{2}/2)].$$We notice the general form of the truncated Morlet wavelet.

## (12)

$${\mathrm{\Psi}}_{M}(\overrightarrow{x})={V}_{\mathrm{\Psi}}(\overrightarrow{x})\mathrm{exp}[i{\varphi}_{\mathrm{\Psi}}(\overrightarrow{x})],\phantom{\rule[-0.0ex]{2em}{0.0ex}}{\varphi}_{\mathrm{\Psi}}(\overrightarrow{x})={k}_{0}x,$$The ridge of the wavelet transform can be computed at each spatial point $\overrightarrow{x}$; it corresponds to a scale ${a}_{r}(\overrightarrow{b})$ such that the local derivative of the wavelet phase ${\varphi}_{\mathrm{\Psi}}$ compensates the local derivative of the object phase $\varphi $.

## (13)

$${f}_{g}\overrightarrow{\nabla}(x,0)+\overrightarrow{\nabla}\varphi (\overrightarrow{x})-\overrightarrow{\nabla}{\varphi}_{\mathrm{\Psi}}[{a}_{r}^{-1}(\overrightarrow{b}){r}_{\theta}(\overrightarrow{x}-\overrightarrow{b})]=0.$$If we consider only the modulated part of the fringe pattern $Q(\overrightarrow{x})\mathrm{exp}[i\varphi (\overrightarrow{x})]$ (complex form), we can derive the equation for its Morlet CWT, given that $Q(\overrightarrow{b})$ changes slowly compared to the phase of the fringes; the rotation angle is fixed, $\theta =0$. It is important to note here that the choice of $\eta =2$ (norm ${\mathcal{L}}^{1}$)^{52}^{,}^{63}^{,}^{64}^{,}^{74} makes this computation straightforward.

## (14)

$${W}_{\mathrm{\Psi}}(\overrightarrow{b},a)=\frac{\widehat{{\mathrm{\Psi}}_{M}}\{a[{f}_{g}(1,0)+\overrightarrow{\nabla}\varphi (\overrightarrow{b})]\}}{C(\overrightarrow{b},a)}Q(\overrightarrow{b}){e}^{i\varphi (\overrightarrow{b})},$$Hence, from Eq. (9), the maxima of the modulus of the CWT correspond to the wavelet ridge skeleton, where the optical phase $\varphi $ produced by the object fulfills the equation

## (15)

$${\overrightarrow{k}}_{0}/{a}_{r}(\overrightarrow{b})={f}_{g}(1,0)+\overrightarrow{\nabla}\varphi (\overrightarrow{b}).$$It can be demonstrated analytically that this ridge detection method is independent of the modulation intensity $Q(\overrightarrow{x})$ [Eq. (2)] of the original fringe pattern (as long as the fringes are detectable) using the properties of Gaussian functions in real and Fourier spaces. This result can be intuitively understood because the ridge detection method boils down to the computation of the position of a local maxima of the wavelet transform with the scale parameter $a$, whatever the value of this maxima. The modulus of the wavelet transform on the ridge skeleton reads

## (16)

$${W}_{\mathrm{\Psi}}[\overrightarrow{b},{a}_{r}(\overrightarrow{b})]=\frac{\widehat{{\mathrm{\Psi}}_{M}}({\overrightarrow{k}}_{0})}{C[\overrightarrow{b},{a}_{r}(\overrightarrow{b})]}Q(\overrightarrow{b}){e}^{i\varphi (\overrightarrow{b})}.$$According to Eq. (16), we could straightforwardly compute $\varphi (\overrightarrow{b})$ from the phase of the wavelet transform ${W}_{\mathrm{\Psi}}[\overrightarrow{b},{a}_{r}(\overrightarrow{b})]$, without using the derivative form of Eq. (15). This is not true when the variations of the phase and their amplitude modulations are too fast compared to ${f}_{g}$, giving a complex value to $C[\overrightarrow{b},{a}_{r}(\overrightarrow{b})]$. In that case, Eq. (15) must be preferred. We also note that the modulus of the wavelet transform on the ridge follows the fringe amplitude modulation $Q(\overrightarrow{b})$. As a general remark, this wavelet-based method intrinsically eliminates background intensity variations $P(\overrightarrow{x})$ that do not affect the fringe pattern modulations.

## 4.

## Validation of the CWT Ridge Detection Method on Latex Microbeads

We discuss the efficiency of the 2-D CWT ridge detection method on a model system made of a microbead particle (radius $R=5\text{\hspace{0.17em}}\text{\hspace{0.17em}}\mu \text{m}$) with refractive index ${n}_{b}=1.59$ surrounded by a matching index oil ${n}_{0}=1.5167$. Figure 3(a) shows the fringe pattern computed with Eq. (2) for the 2-D phase of the bead.

## (17)

$$\varphi (\overrightarrow{x})=[4\pi ({n}_{b}-{n}_{0})\sqrt{{R}^{2}-{(\overrightarrow{x}-{\overrightarrow{x}}_{c})}^{2}}]/\lambda ,$$• 1. The first method uses the 2-D Morlet wavelet as a Fourier filter, with a fixed scale $a=1$ corresponding to the grating fringe modulation ${f}_{g}$; it does not use the ridge detection. We will call it the Fourier filtering method 1.

• 2. The second method uses the 2-D CWT ridge detection method to compute the phase derivative of the fringe pattern, described by Eq. (15), and makes an integration of this derivative along $X$. We will call it the ridge integral method 2.

• 3. The third method uses the 2-D CWT ridge detection method to compute the new complex quantity ${W}_{\mathrm{\Psi}}[b,{a}_{r}(\overrightarrow{b})]$ on the ridge and takes its argument to compute the phase [Eq. (16)]. We will call it the ridge argument method 3.

Figure 4 compares these three methods for phase retrieval on the bead model, for two values of the anisotropy factor $\u03f5$. Figures 4(a) and 4(b) show the phase extracted with a simple Fourier filtering of the fringe pattern, using a Morlet wavelet at fixed scale $a$. This method does not succeed to recover the theoretical phase whatever $\u03f5$ because on the border of the bead, the local frequency of the fringes is too far from the fringe carrier ${f}_{g}$, and it is, therefore, impossible to estimate their phase correctly. Figures 4(c) and 4(d) show the ridge integral method 2 on this bead model. In that case, the theoretical phase can be estimated correctly on the central part of the bead, but computation errors remain on the top and bottom borders of the bead, which come from the difficulty to capture the local CWT modulus maxima position when this modulus vanishes. With method 2, the phase is computed by integration of the gradient $\overrightarrow{\nabla}\varphi (\overrightarrow{b})$ along the axis ${b}_{x}$, which produces an accumulation of the errors on $\varphi (\overrightarrow{x})$ on the right side of the bead. However, switching from isotropic [Fig. 4(c) and $\u03f5=1$] to anisotropic wavelet [Fig. 4(d) and $\u03f5=10$] corrects the errors on the top and the bottom borders of the bead, but the integration errors remain. To improve this aspect, a higher resolution in the modulus maxima detection method is necessary at the expense of computation time. Note, however, that if the fringe pattern is very noisy, method 2 avoids the unwrapping of the phase and may stabilize the computation of the phase. Figures 4(e) and 4(f) show the ridge argument method 3. This method is still sensitive to fringe discontinuities [Fig. 4(e)] on the top and the bottom borders of the bead. With an anisotropic wavelet ($\u03f5=10$), this method succeeds to reproduce the theoretical phase with a ${10}^{-3}$ relative error on the phase. The possibility to adjust the anisotropy of the wavelet is, therefore, important for analyzing phase discontinuities; it has never been reported in that context.

To compare with the bead model, we have imaged a 5 *μ*m radius latex microbead with the QPM and applied the 2-D anisotropic CWT ($\u03f5=10$) to analyze the experimental fringe pattern shown in Fig. 5(a). Unlike the theoretical model, the fringe intensity modulation in the QPM is no longer constant, see the horizontal section plotted in black in Fig. 5(b). This intensity modulation is produced by the light scattering by the bead. As established by Bedrosian,^{76} when this modulation contains frequencies that overlap with the fringe carrier frequency ${f}_{g}$, the extraction of the phase with the Hilbert method is biased and leads to a false estimation. Indeed, from the Hilbert transform of the term $Q(\overrightarrow{x})\mathrm{cos}[{f}_{g}x+\varphi (\overrightarrow{x})]$ in Eq. (2), we would like to recover a function like $Q(\overrightarrow{x})\mathrm{sin}[{f}_{g}x+\varphi (\overrightarrow{x})]$ from which the phase could be computed straightforwardly. Bedrosian’s theorem shows that this is workable only if the amplitude modulation $Q(\overrightarrow{x})$ does not contain frequencies that mix with the carrier frequency ${f}_{g}$. This condition is not experimentally satisfied as illustrated in Fig. 5(b). In that case, when using the ridge detection, the CWT offers the advantage of leading to a measure of the frequency from the ridge that considerably reduces the effects of fringe modulation amplitude on phase retrieval [Eq. (15)]. In Fig. 5(c), we show the modulus of the 2-D CWT $|{W}_{\mathrm{\Psi}}({b}_{x},a)|$ on the horizontal section shown in Fig. 5(a). The ridge ${a}_{r}(\overrightarrow{b})$ plotted in Fig. 5(c) with a black line is very similar to the theoretical curve shown in Fig. 3(c). The real part of the CWT computed on the ridge shown in Fig. 5(d) is very impressive when compared to the original image [Fig. 5(a)]. It looks as if the major intensity modulations of the original images were damped out to keep only the fringe distorsion, from which the phase is computed. The same section of Fig. 5(d) is plotted in gray in Fig. 5(b). The efficiency of the wavelet transform is well illustrated in this example. Figure 5(e) shows the modulus $|{W}_{\mathrm{\Psi}}[\overrightarrow{b},{a}_{r}(\overrightarrow{b})]|$ computed on the ridge of the 2-D CWT on this latex microbead; it is coded in gray from minimum (black) to maximum (white). Figure 5(f) gives a 2-D gray-coded image of the phase derivative $\partial \varphi (\overrightarrow{b})/\partial b$ of the CWT on the ridge, and Fig. 5(g) shows the three-dimensional (3-D) representation of the phase $\varphi (\overrightarrow{x})$ computed from the ridge of the CWT with method 3. The modulus of the 2-D CWT on the ridge preserves the intensity of the original fringe image, whereas this intensity modulation disappears completely on the phase derivative $\partial \varphi (\overrightarrow{b})/\partial b$. We also note that as for the bead model, the anisotropic Morlet wavelet allows a very nice detection of the phase derivative. The 3-D picture of the phase in Fig. 5(g) is very appealing since it not only detects perfectly the phase drop due to this bead and reproduces the theoretical prediction, but also delineates some defects of the beads that are visible as phase irregularities on the surface of the bead. We have estimated the standard deviation of the background phase on this image of $\sim 0.25\text{\hspace{0.17em}}\text{\hspace{0.17em}}\mathrm{rad}$.

## 5.

## Application of the CWT Phase Retrieval Method to Living Cell Imaging

We illustrate the performance of the CWT phase retrieval method on a murine myoblast cell line C2C12, which can be differentiated to give rise to plurinucleate syncytia (the myotubes) by fusion. We will focus here on undifferentiated C2C12 cell monolayers. Figure 6 reports the 2-D CWT ridge analysis of a QPM image of a nonadherent myoblast that is progressively rounding and will detach from the glass to enter mitosis.^{77} The characteristic organization of the microtubule-actin cytoskeleton with stress fibers has disappeared, the cell is round, and it is no longer possible to delineate a nice nuclear contour inside this cell. However, this spherical shape is interesting because it allows us to perform the same parametrization of the phase map as we have done previously for spherical particles. Compared to the previous example of the microbead in Sec. 4, we observe on Fig. 6(a) that besides scattering effects on the border of the bead, the internal structure of the cell also produces intensity modulations of the fringes that may make the analysis more complex. We have plotted in Fig. 6(b) a profile of the fringe image selected on the white dashed section ofFig. 6(a) (black curve), which can be compared to the corresponding profile of the real part of the 2-D CWT ridge on the same section. Similar to the example of Fig. 5, the CWT ridge analysis regularizes the fringe intensity modulation. Figure 6(c) reports the modulus of the 2-D CWT $|{W}_{\mathrm{\Psi}}({b}_{x},a)|$ on the horizontal section shown in Fig. 6(a). The ridge ${a}_{r}(\overrightarrow{b})$ plotted in Fig. 6(c) with a black line is reminiscent of the one shown in Fig. 5(c). The real part of the CWT computed on the ridge shown in Fig. 6(d) when compared to the original QPM image [Fig. 6(a)] definitely flattens the intensity modulations. Figure 6(e) shows the modulus $|{W}_{\mathrm{\Psi}}[\overrightarrow{b},{a}_{r}(\overrightarrow{b})]|$ computed on the ridge of 2-D CWT; it is coded in gray from minimum (black) to maximum (white). The intensity modulation that remains in Fig. 6(e) is eliminated in the derivative of the phase shown in Fig. 6(f) (ridge method 2). Finally, using method 3, we reconstruct a 3-D profile of the phase of this myoblast in Fig. 6(g). This representation confirms the round shape of this cell in the third direction $Z$ and points out a central part where smaller objects are identified, which may be attributed to condensed chromosomes. We use the same model as proposed for a spherical bead as a first approximation to extract the mean refractive index of this cell and delineate one of the protruding object and its refractive index. The interpolation of the whole cell by a spherical phase shape [Eq. (17)] is reported in Fig. 7. This interpolation leads to an estimation of the overall refractive index of the cell as ${n}_{c}=1.36\pm 0.005$, given that the buffer refractive index is ${n}_{0}=1.33$. Moreover, subtracting this mean spherical phase contour from the total phase contour (see Fig. 7), we can also compute the phase drop of the small spherical object that pops up on the upper part of the cell and estimate its index to be $1.4\pm 0.005$.

In Fig. 8, we show the image of an adherent myoblast cell, where we can recognize thin lamellipodia extensions that exhibit filopodia projections on the leading edge: a characteristic pattern of cell motility. It has been shown in the literature that differentiation and fusion of myoblasts into multinucleated myotubes is accompanied by a dramatic reorganization of the Golgi complex.^{78} Here we rather have the classic compact juxtanuclear Golgi complex of an undifferentiated myoblast that we can recognize as small granular objects on both the derivative of the phase $\varphi $ reported in Fig. 8(c) and on the phase $\varphi $ shown in a 3-D representation in Fig. 8(d). The phase response of this adherent cell is different from that shown for the round cell [Fig. 6(g)]; the nucleus looks more like a phase plateau, flattened by the mechanical tractions of the lamellipodia. On the border of the nucleus, the QPM detects a necklace of phase droplets and a central part with higher phase [Figs. 8(c) and 8(d)]. At this stage, it is difficult to conclude if these small bodies are essentially Golgi complex or a combination of perinuclear organelles as rough and smooth endoplasmic reticulum, Golgi, vesicles, and mitochondrial network.

## 6.

## Summary

We have shown that a wavelet-based space-scale analysis can be used to decode the fringe images recorded from living cells with a QPM. The implementation of the ridge detection method is more successful than Fourier filtering methods when imaging cellular and high refractive index contrast objects since it can discriminate intensity from phase changes. This technique has been applied to undifferentiated myoblast cells and revealed internal structures of these cells, which were confirmed by fluorescence imaging. This microscope is coupled to a high-speed camera, and we hope in the near future that besides capturing the optical phase changes produced by the cell internal structures, the record of the dynamics of these internal bodies will provide a complementary way to distinguish these structures without the need of fluorescence staining.

## 7.

## Materials and Methods

## 7.1.

### Quantitative Phase Microscope

A low-coherence laser diode (Thorlabs, GmbH, Germany, $\lambda =532\text{\hspace{0.17em}}\text{\hspace{0.17em}}\mathrm{nm}$) is used as a light source and is directed to the sample (S) using Köhler illumination, such that the field at the image plane (IP) is spatially coherent over the entire field of view (Fig. 1). Different diffraction orders are then created with a transmission grating (G) ($70\text{\hspace{0.17em}}\text{\hspace{0.17em}}\mathrm{grooves}/\mathrm{mm}$) localized at the IP of the microscope equipped with an objective (O) $40\times $ (Olympus, France, SPlan40, $\mathrm{N}.\mathrm{A}.=0.7$). A spatial filter (Thorlabs, custom-made) is placed at the Fourier plane of lens ${L}_{1}$ to select the first-order beam (imaging field) and to low-pass filter the zeroth-order beam (reference). The spatial filter has been designed with two circular apertures with diameters of 15 *μ*m and 2 mm. The two beams are recombined using a second Fourier lens (${L}_{2}$), and the resulting interferogram is recorded as an image of $2048\times 2048$ pixels with a CMOS camera (Hamamatsu, Japan, ORCA-Flash 4.0). The $4f$ lens system adds a $5\times $ magnification (${f}_{1}=50\text{\hspace{0.17em}}\text{\hspace{0.17em}}\mathrm{nm}$, ${f}_{2}=250\text{\hspace{0.17em}}\text{\hspace{0.17em}}\mathrm{mm}$).

## 7.2.

### Polymer Layer Preparation

A solution of 10% poly(methyl methacrylate) (PMMA, Sigma-Aldrich, France) in toluene was spin coated over a glass coverslip at a speed of 1000 rpm during 50 s. A time of 10 s is fixed to reach the nominal speed. After coating, the film was annealed for 1 h at 140°C. Prior to imaging, a portion of the PMMA layer was removed with a scalpel.

## 7.3.

### Polystyrene Beads Preparation

1 *μ*L of an aqueous solution containing polystyrene beads (FLUKA 72986) was diluted in 10 mL of deionized water. 250 *μ*L of the dilution were deposited on a small petri dish with a glass window and let overnight in an oven at 70°C to dry. Before imaging, 500 *μ*L of glass index matching oil were added.

## 7.4.

### Cell Culture

C2C12 mouse cells (ATCC number CRL-1772™) were grown in high glucose ($4.5\text{\hspace{0.17em}}\text{\hspace{0.17em}}\mathrm{g}/\mathrm{L}$) Dulbecco’s modified Eagle medium [(DMEM), GE Healthcare Life Science, Dominique Dutscher, France] supplemented with 20% fetal bovine serum GE Healthcare Life Science and 1% antibiotics (penicillin/streptamicine). Adherent myoblasts on 50 mm petri dishes with a glass bottom of 0.17 mm thickness were maintained at 37°C and 5% ${\mathrm{CO}}_{2}$ up to 60% confluence until they were used. The glass surface was not treated to enhance cell adhesion. The growth medium was replaced by phosphate buffered saline solution before imaging. No further sample preparation was required.

## Acknowledgments

We are indebted to Centre National de la Recherche Scientifique, Ecole Normale Supérieure de Lyon, Lyon Science Transfert (projet L659), Région Rhône Alpes (CIBLE Program 2011), INSERM (AAP Physique Cancer 2012), and the French Agency for Research (ANR-AA-PPPP-005, EMMA 2011) for their financial support. C.M.T. is very grateful to the National Council of Science and Technology (CONACyT, Mexico) for supporting her PhD scholarship. L.S. is very thankful to the Association Française Contre les Myopathies, AFM-Téléthon, for supporting her PhD scholarship.

## References

M. BornE. Wolf, Principles of Optics: Electromagnetic Theory of Propagation, Interference and Diffraction of Light, 7th ed., Cambridge University Press, Cambridge (1999).Google Scholar

F. Zernike, “How I discovered phase contrast,” Science 121(3141), 345–349 (1955).SCIEAS0036-8075http://dx.doi.org/10.1126/science.121.3141.345Google Scholar

G. NomarskiA. R. Weill, “Application à la métallographie des méthodes interférentielles à deux ondes polarisées,” Revue de Métallurgie 2, 121–128 (1955).Google Scholar

A. D. Gabor, “Microscopy by reconstructed wave-fronts,” Proc. R. Soc. Lon. A 197(1051), 454–487 (1949).PEAMDU0308-2105http://dx.doi.org/10.1098/rspa.1949.0075Google Scholar

B. Rappazet al., “Measurement of the integral refractive index and dynamic cell morphometry of living cells with digital holographic microscopy,” Opt. Express 13(23), 9361–9373 (2005).OPEXFF1094-4087http://dx.doi.org/10.1364/OPEX.13.009361Google Scholar

L. Yuet al., “Digital holographic microscopy for quantitative cell dynamic evaluation during laser microsurgery,” Opt. Express 17(14), 12031–12038 (2009).OPEXFF1094-4087http://dx.doi.org/10.1364/OE.17.012031Google Scholar

V. P. TychinskyA. N. Tikhonov, “Interference microscopy in cell biophysics. 1. Principles and methodological aspects of coherent phase microscopy,” Cell Biochem. Biophys. 58(3), 107–116 (2010).CBBIFV1085-9195http://dx.doi.org/10.1007/s12013-010-9114-zGoogle Scholar

G. Popescu, Quantitative Phase Imaging of Cells and Tissues, McGraw Hill, New York (2011).Google Scholar

N. CardenasS. KumarS. Mohanty, “Dynamics of cellular response to hypotonic stimulation revealed by quantitative phase microscopy and multi-fractal detrended fluctuation analysis,” Appl. Phys. Lett. 101(20), 203702 (2012).APPLAB0003-6951http://dx.doi.org/10.1063/1.4766922Google Scholar

N. T. Shaked, “Quantitative phase microscopy of biological samples using a portable interferometer,” Opt. Lett. 37(11), 2016–2018 (2012).OPLEDP0146-9592http://dx.doi.org/10.1364/OL.37.002016Google Scholar

N. Sugiyamaet al., “Label-free characterization of living human induced pluripotent stem cells by subcellular topographic imaging technique using full-field quantitative phase microscopy coupled with interference reflection microscopy,” Biomed. Opt. Express 3(9), 2175–2183 (2012).BOEICL2156-7085http://dx.doi.org/10.1364/BOE.3.002175Google Scholar

P. BonB. WattellierS. Monneret, “Modeling quantitative phase image formation under tilted illuminations,” Opt. Lett. 37(10), 1718–1720 (2012).OPLEDP0146-9592http://dx.doi.org/10.1364/OL.37.001718Google Scholar

P. Bonet al., “Optical detection and measurement of living cell morphometric features with single-shot quantitative phase microscopy,” J. Biomed. Opt. 17(7), 076004 (2012).JBOPFO1083-3668http://dx.doi.org/10.1117/1.JBO.17.7.076004Google Scholar

V. Tychinsky, “The metabolic component of cellular refractivity and its importance for optical cytometry,” J. Biophotonics 2(8–9), 494–504 (2009).JBOIBX1864-063Xhttp://dx.doi.org/10.1002/jbio.v2:8/9Google Scholar

Y. Parket al., “Measurement of red blood cell mechanics during morphological changes,” Proc. Natl. Acad. Sci. U. S. A. 107(15), 6731–6836 (2010).PNASA60027-8424http://dx.doi.org/10.1073/pnas.0909533107Google Scholar

Y. Parket al., “Metabolic remodeling of the human red blood cell membrane,” Proc. Natl. Acad. Sci. U. S. A. 107(4), 1289–1294 (2010).PNASA60027-8424http://dx.doi.org/10.1073/pnas.0910785107Google Scholar

T. E. GureyevA. RobertsK. A. Nugent, “Phase retrieval with the transport-of-intensity equation: matrix solution with use of Zernike polynomials,” J. Opt. Soc. Am. A 12(9), 1932–1941 (1995).JOAOD60740-3232http://dx.doi.org/10.1364/JOSAA.12.001932Google Scholar

T. E. GureyevA. RobertsK. A. Nugent, “Partially coherent fields, the transport-of-intensity equation, and phase uniqueness,” J. Opt. Soc. Am. A 12(9), 1942–1946 (1995).JOAOD60740-3232http://dx.doi.org/10.1364/JOSAA.12.001942Google Scholar

S. Bajtet al., “Quantitative phase-sensitive imaging in a transmission electron microscope,” Ultramicroscopy 83(1–2), 67–73 (2000).ULTRD60304-3991http://dx.doi.org/10.1016/S0304-3991(99)00174-6Google Scholar

M. R. Teague, “Image formation in terms of the transport equation,” J. Opt. Soc. Am. A 2(11), 2019–2026 (1985).JOAOD60740-3232http://dx.doi.org/10.1364/JOSAA.2.002019Google Scholar

G. Ade, “On the validity of the transport equation for the intensity in optics,” Opt. Commun. 52(5), 307–310 (1985).OPCOB80030-4018http://dx.doi.org/10.1016/0030-4018(85)90190-7Google Scholar

M. Beleggiaet al., “On the transport of intensity technique for phase retrieval,” Ultramicroscopy 102(1), 37–49 (2004).ULTRD60304-3991http://dx.doi.org/10.1016/j.ultramic.2004.08.004Google Scholar

A. Bartyet al., “Quantitative optical phase microscopy,” Opt. Lett. 23(11), 817–819 (1998).OPLEDP0146-9592http://dx.doi.org/10.1364/OL.23.000817Google Scholar

C. J. Bellairet al., “Quantitative phase amplitude microscopy IV: imaging thick specimens,” J. Microsc. 214(Pt 1), 62–69 (2004).JMICAR0022-2720http://dx.doi.org/10.1111/j.0022-2720.2004.01302.xGoogle Scholar

G. Popescuet al., “Diffraction phase microscopy for quantifying cell structure and dynamics,” Opt. Lett. 31(6), 775–777 (2006).OPLEDP0146-9592http://dx.doi.org/10.1364/OL.31.000775Google Scholar

Y. Parket al., “Diffraction phase and fluorescence microscopy,” Opt. Express 14(18), 8263–8268 (2006).OPEXFF1094-4087http://dx.doi.org/10.1364/OE.14.008263Google Scholar

B. Bhaduriet al., “Diffraction phase microscopy with white light,” Opt. Lett. 37(6), 1094–1096 (2012).OPLEDP0146-9592http://dx.doi.org/10.1364/OL.37.001094Google Scholar

T. Ikedaet al., “Hilbert phase microscopy for investigating fast dynamics in transparent systems,” Opt. Lett. 30(10), 1165–1167 (2005).OPLEDP0146-9592http://dx.doi.org/10.1364/OL.30.001165Google Scholar

H. V. Phamet al., “Real time blood testing using quantitative phase imaging,” PloS One 8(2), e55676 (2013).1932-6203http://dx.doi.org/10.1371/journal.pone.0055676Google Scholar

B. BhaduriG. Popescu, “Derivative method for phase retrieval in off-axis quantitative phase imaging,” Opt. Lett. 37(11), 1868–1870 (2012).OPLEDP0146-9592http://dx.doi.org/10.1364/OL.37.001868Google Scholar

H. V. Phamet al., “Fast phase reconstruction in white light diffraction phase microscopy,” Appl. Opt. 52(1), A97–101 (2013).APOPAI0003-6935http://dx.doi.org/10.1364/AO.52.000A97Google Scholar

J. Morletet al., “Wave propagation and sampling theory—Part II: Sampling theory and complex waves,” Geophysics 47(2), 222–236 (1982).GPYSA70016-8033http://dx.doi.org/10.1190/1.1441329Google Scholar

M. TakedaH. InaS. Kobayashi, “Fourier-transform method of fringe-pattern analysis for computer-based topography and interferometry,” J. Opt. Soc. Am. 72(1), 156–160 (1982).JOSAAH0030-3941http://dx.doi.org/10.1364/JOSA.72.000156Google Scholar

C. RoddierF. Roddier, “Interferogram analysis using Fourier transform techniques,” Appl. Opt. 26(9), 1668–16673 (1987).APOPAI0003-6935http://dx.doi.org/10.1364/AO.26.001668Google Scholar

P. Flandrin, Time-Frequency / Time-Scale Analysis, Academic Press, New York (1999).Google Scholar

P. GoupillaudA. GrossmannJ. Morlet, “Cycle-octave and related transforms in seismic signal analysis,” Geoexploration 23(1), 85–102 (1984).GEOXAV0016-7142http://dx.doi.org/10.1016/0016-7142(84)90025-5Google Scholar

A. GrossmannJ. Morlet, “Decomposition of Hardy functions into square integrable wavelets of constant shape,” SIAM J. Math. Anal. 15(4), 723–736 (1984).SJMAAH0036-1410http://dx.doi.org/10.1137/0515056Google Scholar

J. M. CombesA. GrossmannP. Tchamitchian, Eds., Wavelets, Springer, Berlin (1989).Google Scholar

I. Daubechies, Ten Lecture on Wavelets, SIAM, Philadelphia (1992).Google Scholar

Y. Meyer, Ed., Wavelets and Their Applications, Springer, Berlin (1992).Google Scholar

M. B. Ruskaiet al., Eds., Wavelets and Their Applications, Jones and Bartlett, Boston (1992).Google Scholar

P. Flandrin, Temps-Fréquence, Hermès, Paris (1993).Google Scholar

Y. MeyerS. Roques, Eds., Progress in Wavelets Analysis and Applications, Editions Frontières, Gif-sur-Yvette (1993).Google Scholar

A. Arneodoet al., Ondelettes, Multifractales et Turbulences: de l’ADN aux croissances cristallines, Diderot Editeur, Art et Sciences, Paris (1995).Google Scholar

A. ArneodoE. BacryJ. F. Muzy, “The thermodynamics of fractals revisited with wavelets,” Physica A 213(1–2), 232–275 (1995).PHYADX0378-4371http://dx.doi.org/10.1016/0378-4371(94)00163-NGoogle Scholar

G. ErlebacherM. Y. HussainiL. M. Jameson, Eds., Wavelets: Theory and Applications, Oxford University Press, Oxford (1996).Google Scholar

S. Mallat, A Wavelet Tour of Signal Processing, Academic Press, New York (1998).Google Scholar

B. Torresani, Analyse Continue par Ondelettes, Editions de Physique, Les Ulis (1998).Google Scholar

A. ArneodoN. DecosterS. Roux, “A wavelet-based method for multifractal image analysis. I. Methodology and test applications on isotropic and anisotropic random rough surfaces,” Eur. Phys. J. B 15(3), 567–600 (2000).EPJBFY1434-6028http://dx.doi.org/10.1007/s100510051161Google Scholar

A. Arneodoet al., “A wavelet-based method for multifractal image analysis: from theoretical concepts to experimental applications,” Adv. Imaging Electron Phys. 126, 1–92 (2003).AIEPFQ1076-5670http://dx.doi.org/10.1016/S1076-5670(03)80014-9Google Scholar

P. KestenerA. Arneodo, “Generalizing the wavelet-based multifractal formalism to random vector fields: application to three-dimensional turbulence velocity and vorticity data,” Phys. Rev. Lett. 93(4), 044501 (2004).PRLTAO0031-9007http://dx.doi.org/10.1103/PhysRevLett.93.044501Google Scholar

J. P. Antoineet al., Two-Dimensional Wavelets and Their Relatives, Cambridge University Press, Cambridge, United Kingdom (2004).Google Scholar

A. ArneodoB. AuditP. Kestener, “Wavelet-based multifractal analysis,” Scholarpedia 3(3), 4103 (2008).1941-6016http://dx.doi.org/10.4249/scholarpedia.4103Google Scholar

A. Arneodoet al., “Multi-scale coding of genomic information: from DNA sequence to genome structure and function,” Phys. Rep. 498(2–3), 45–188 (2011).PRPLCM0370-1573http://dx.doi.org/10.1016/j.physrep.2010.10.001Google Scholar

E. Freyszet al., “Optical wavelet transform of fractal aggregates,” Phys. Rev. Lett. 64(7), 745–748 (1990).PRLTAO0031-9007http://dx.doi.org/10.1103/PhysRevLett.64.745Google Scholar

J. F. Muzyet al., “Optical-diffraction measurement of fractal dimensions and $f(\alpha )$ spectrum,” Phys. Rev. A 45(12), 8961–8964 (1992).PLRAAN1050-2947http://dx.doi.org/10.1103/PhysRevA.45.8961Google Scholar

W. W. Macy, “Two-dimensional fringe-pattern analysis,” Appl. Opt. 22(23), 3898–3901 (1983).APOPAI0003-6935Google Scholar

P. Sandoz, “Wavelet transform as a processing tool in white-light interferometry,” Opt. Lett. 22(14), 1065–1067 (1997).OPLEDP0146-9592http://dx.doi.org/10.1364/OL.22.001065Google Scholar

P. Tomassiniet al., “Analyzing laser plasma interferograms with a continuous wavelet transform ridge extraction technique: the method,” Appl. Opt. 40(35), 6561–6568 (2001).APOPAI0003-6935http://dx.doi.org/10.1364/AO.40.006561Google Scholar

J. ZhongJ. Weng, “Spatial carrier-fringe pattern analysis by means of wavelet transform: wavelet transform profilometry,” Appl. Opt. 43(26), 4993–4998 (2004).APOPAI0003-6935http://dx.doi.org/10.1364/AO.43.004993Google Scholar

H. Liu, “Experimental verification of improvement of phase shifting Moiré interferometry using wavelet-based image processing,” Opt. Eng. 43(5), 1206–1214 (2004).OPEGAR0091-3286http://dx.doi.org/10.1117/1.1690281Google Scholar

J. ZhongJ. Weng, “Phase retrieval of optical fringe pattern from the ridge of a wavelet transform,” Opt. Lett. 30(19), 2560–2562 (2005).OPLEDP0146-9592http://dx.doi.org/10.1364/OL.30.002560Google Scholar

Z. WangH. Ma, “Advanced continuous wavelet transform algorithm for digital interferogram analysis and processing,” Opt. Eng. 45(4), 045601 (2006).OPEGAR0091-3286http://dx.doi.org/10.1117/1.2188399Google Scholar

S. LiX. SuW. Chen, “Wavelet ridge techniques in optical fringe pattern analysis,” J. Opt. Soc. Am. A 27(6), 1245–1254 (2010).JOAOD60740-3232http://dx.doi.org/10.1364/JOSAA.27.001245Google Scholar

M. A. Gdeisatet al., “Spatial and temporal carrier fringe pattern demodulation using the one-dimensional continuous wavelet transform: recent progress, challenges, and suggested developments,” Opt. Lasers Eng. 47(12), 1348–1361 (2009).OLENDN0143-8166http://dx.doi.org/10.1016/j.optlaseng.2009.07.009Google Scholar

R. Oven, “Extraction of phase derivative data from interferometer images using a continuous wavelet transform to determine two-dimensional refractive index profiles,” Appl. Opt. 49(22), 4228–4236 (2010).APOPAI0003-6935http://dx.doi.org/10.1364/AO.49.004228Google Scholar

Z. Zhanget al., “Comparison of Fourier transform, windowed Fourier transform, and wavelet transform methods for phase calculation at discontinuities in fringe projection profilometry,” Opt. Lasers Eng. 50(8), 1152–1160 (2012).OLENDN0143-8166http://dx.doi.org/10.1016/j.optlaseng.2012.03.004Google Scholar

M. A. GdeisatD. R. BurtonM. J. Lalor, “Spatial carrier fringe pattern demodulation by use of a two-dimensional continuous wavelet transform,” Appl. Opt. 45(34), 8722–8732 (2006).APOPAI0003-6935http://dx.doi.org/10.1364/AO.45.008722Google Scholar

Q. Kemao, “Two-dimensional windowed Fourier transform for fringe pattern analysis: principles, applications and implementations,” Opt. Lasers Eng. 45(2), 304–317 (2007).OLENDN0143-8166http://dx.doi.org/10.1016/j.optlaseng.2005.10.012Google Scholar

A. FedericoG. H. Kaufmann, “Phase retrieval of singular scalar light fields using a two-dimensional directional wavelet transform and a spatial carrier,” Appl. Opt. 47(28), 5201–5207 (2008).APOPAI0003-6935http://dx.doi.org/10.1364/AO.47.005201Google Scholar

H. NiuC. QuanC. Tay, “Phase retrieval of speckle fringe pattern with carriers using 2D wavelet transform,” Opt. Lasers Eng. 47(12), 1334–1339 (2009).OLENDN0143-8166http://dx.doi.org/10.1016/j.optlaseng.2008.10.005Google Scholar

S. LiX. SuW. Chen, “Spatial carrier fringe pattern phase demodulation by use of a two-dimensional real wavelet,” Appl. Opt. 48(36), 6893–6906 (2009).APOPAI0003-6935http://dx.doi.org/10.1364/AO.48.006893Google Scholar

K. PokorskiK. Patorski, “Visualization of additive-type Moiré and time-average fringe patterns using the continuous wavelet transform,” Appl. Opt. 49(19), 3640–3651 (2010).APOPAI0003-6935http://dx.doi.org/10.1364/AO.49.003640Google Scholar

J. Maet al., “Two-dimensional continuous wavelet transform for phase determination of complex interferograms,” Appl. Opt. 50(16), 2425–2430 (2011).APOPAI0003-6935http://dx.doi.org/10.1364/AO.50.002425Google Scholar

K. PokorskiK. Patorski, “Separation of complex fringe patterns using two-dimensional continuous wavelet transform,” Appl. Opt. 51(35), 8433–8439 (2012).APOPAI0003-6935http://dx.doi.org/10.1364/AO.51.008433Google Scholar

E. Bedrosian, “Product theorem for Hilbert transforms,” Proc. IEEE 51, 868–869 (1963).IEEPAD0018-9219http://dx.doi.org/10.1109/PROC.1963.2308Google Scholar

B. StädlerT. M. BlättlerA. Franco-Obregón, “Time-lapse imaging of in vitro myogenesis using atomic force microscopy,” J. Microsc. 237(1), 63–69 (2010).JMICAR0022-2720http://dx.doi.org/10.1111/(ISSN)1365-2818Google Scholar

E. Ralston, “Changes in architecture of the Golgi complex and other subcellular organelles during myogenesis.” J. Cell Biol. 120(2), 399–409 (1993).JCLBA30021-9525http://dx.doi.org/10.1083/jcb.120.2.399Google Scholar

## Biography

**Cristina Martinez-Torres** obtained a BSc in biophysics from the Autonomous University of San Luis Potosi, Mexico. During her undergraduate studies, she was involved in several research groups, working in topics such as bioinformatics, cellular biology, biophysics, optical engineering, and atomic physics. Currently, she is a PhD student in the group of Françoise Argoul at the Ecole Normale Suprieure de Lyon, France, where she studies cell dynamics.

**Lotfi Berguiga** received his PhD in physics from the University of Burgundy in France in 2001, where he worked in near field optics. Since 2004, he has worked at the Ecole Normale Suprieure de Lyon, where he has investigated surface plasmon microscopy. Since 2007, he has worked as research engineer at the French National Centre for Scientific Research, and his research interests include development of new techniques of microscopy applied to cell biology.

**Laura Streppa** obtained her MSc with honors in biomolecular research from University of Urbino “Carlo Bo” and performed three years compulsory and volunteer internship in different laboratories (microbiology, molecular biology, biochemistry, and physics). She is currently a first-year PhD student at PHAST of Lyon. Her thesis, titled “Biomechanical study of neuromuscular junction and its impact in myopathies,” is supported by AFM Telethon and is a collaboration between Laboratory of Physics and LMBC of ENS.

**Elise Boyer-Provera** graduated in optics and industrial visual systems from the Engineering School of Telecom in Saint Etienne in 2010. After a three-year training in the private company Thales Angenieux, she was hired as an engineer at Ecole Normale Suprieure de Lyon to develop original microscopy devices based on surface plasmon microscopy and quantitative phase microscopy.

**Laurent Schaeffer** obtained his PhD thesis in molecular biology in 1995 from the University of Strasbourg, France. He is presently the head of the Center of Cell Biotechnology of Lyon Hospitals and the director of the Laboratory of Molecular Biology of the Cell in ENS Lyon. He currently studies the mechanisms that control muscle and neuromuscular junction formation and maintenance with a special focus on the control of muscle gene expression by neural cues.

**Juan Elezgaray** is a senior researcher at CNRS, in the unit CBMN (UMR 5248, Pessac, France). He received his PhD in theoretical physics from Paris 6 University in 1989. He has held a permanent position at CNRS since 1989. His scientific interests are focused on modeling biologically motivated systems, from proteins to DNA; he also develops optics-related modeling.

**Alain Arneodo** got his thesis in elementary particle physics from the University of Nice, France, in 1978. He is the director of research at CNRS and has published over 200 peer-reviewed papers. His scientific contribution encompasses many fields of modern physics, including statistical mechanics, dynamical systems theory, fully developed turbulence, the mathematics of fractals and multifractals, signal and image processing, wavelet transform analysis, and its applications in physics, geophysics, astrophysics, chemistry, biology, and finance.

**Françoise Argoul** received her PhD from the University of Bordeaux in 1986. She is director of research at CNRS and has coauthored over 110 peer-reviewed papers. She manages a research program in Laboratory of Physics of Ecole Normale Superieure of Lyon, devoted to the experimental characterization of the mechanical and genomic response of cellular systems under external stress and to the role of mechanical and dynamical parameters on the modification of cellular mechanims.