Methylene-blue aided rapid confocal laser endomicroscopy of breast cancer

Breast conserving surgery allows complete tumor resection while maintaining acceptable cosmesis for patients. Safe and rapid intraoperative margin assessment during the procedure is important to establish the completeness of tumor excision and minimizes the need for reoperation. Confocal laser endomicroscopy has demonstrated promise for real-time intraoperative margin assessment using acriflavine staining, but it is not approved for routine in-human use. We describe a custom high-speed line-scan confocal laser endomicroscopy (LS-CLE) system at 660 nm that enables high-resolution histomorphological imaging of breast tissue stained with methylene-blue, an alternative fluorescent stain for localizing sentinel nodes during breast surgery. Preliminary imaging results on freshly excised human breast tissue specimens are presented, demonstrating the potential of methylene-blue aided rapid LS-CLE to determine the oncological status of surgical margins in-vivo. © The Authors. Published by SPIE under a Creative Commons Attribution 3.0 Unported License. Distribution or reproduction of this work in whole or in part requires full attribution of the original publication, including its DOI. [DOI: 10.1117/1.JBO.22.2.020501]

Breast cancer is a major public health problem worldwide and the second leading cause of cancer-related female deaths.In 2012, there were an estimated 464,000 new cases of female breast cancer and 131,000 deaths in Europe alone. 1Breast conserving surgery (BCS), in the form of wide local excision with glandular remodeling, is an attractive option for patients as it allows complete tumor resection while maintaining acceptable cosmesis and the oncological outcome is similar to a mastectomy. 2,3However, on average, ∼20% to 30% of BCS patients require one or more reoperative interventions, mainly due to the presence of positive or close margins in postoperative histological analysis of the resected tissue. 4,5r breast surgical procedures, conventional excision biopsy followed by histology imaging is still required for comprehensive assessment of margin status.While traditional histology provides high diagnostic accuracy, it is labor-intensive and slow, adding ∼20 to 30 min to the duration of the operation. 6ne potential avenue for reducing reoperation rates following BCS is the use of high-resolution intraoperative imaging to determine if the radial margins are tumor-free during surgery.Optical imaging techniques, including optical coherence tomography, 7 confocal reflectance microscopy, 8 and nonlinear microscopy, 9 are currently being investigated for in-vivo visualization of breast morphology.
Probe-based confocal laser endomicroscopy (pCLE) is one such emerging optical biopsy technique that is suitable for intraoperative margin assessment as it can provide real-time imaging of tissue structure with microscopic resolution.The method typically uses a flexible optical fiber bundle to relay light to and from the tissue, and cellular features are made visible by exogenous staining prior to imaging.A preliminary study on 71 ex-vivo breast tissue samples imaged with the Cellvizio pCLE system (Mauna Kea Technologies) demonstrated that cancerous and normal tissue could be differentiated by histopathologists with an accuracy of 94%. 10 This study used topically applied acriflavine, excited at 488 nm, which stains cell nuclei and enables high contrast imaging of tissue morphology.However, this is not currently an attractive option for in-vivo imaging due to the lack of regulatory approval and suggested risks of mutagenicity. 11o avoid these safety concerns, a subsequent in-vivo human study used intravenous fluorescein as the contrast agent, a compound that is well-established as being safe for in-vivo human use 12 and is commonly employed for other clinical applications of pCLE. 13,14However, since fluorescein staining is nonspecific, it does not highlight tissue morphology as effectively, causing difficulties in distinguishing between benign and malignant lesions. 12or intraoperative margin assessment in routine clinical practice, it is desirable to use endomicroscopes with approved fluorescent dyes, which are both safe for in-vivo human use and provide sufficient contrast to discriminate between different tissue morphologies.In this letter, we demonstrate that normal and cancerous breast tissue stained with methylene-blue, and excited at 660 nm, exhibits comparable morphological features to acriflavine-stained tissue.6][17] Topical methyelene blue has also been demonstrated to image precancerous and neoplastic lesions of central airways using a CLE system. 18,19mages were acquired ex-vivo using a custom high-speed line-scan confocal laser endomicroscopy (LS-CLE) system operating at 660 nm.The system configuration is illustrated in Fig. 1(a).The system set-up is similar to a previously developed virtual slit LS-CLE at 488 nm but is adapted and optimized for methylene-blue imaging.A full description of the system's operating principles can be found in Ref. 20.In brief, a cylindrical lens (f ¼ 50 mm) is used to create a focused line from a 100 mW, 660 nm laser (Vortran Stradus, 660).A galvomirror (Thorlabs GVS001) sweeps the line across the proximal end of the fiber bundle probe (Cellvizio Gastroflex UHD, Mauna Kea Technologies) in a direction perpendicular to the line.
The fiber bundle probe relays the line to the tissue via a distal objective and returns collected fluorescence from all the points along the line, which is then imaged onto a monochrome rollingshutter CMOS camera (Flea 3, FL3-U3-13S2M-CS).In this design, the rolling shutter of the CMOS camera operates as a virtual detector slit that rejects most of the out-of-focus light, leading to optical sectioning at frame rates of up to 120 Hz.To accommodate the 660-nm excitation wavelength, the dichroic and emission filters from the system reported previously 20 were replaced with a Thorlabs-DMLP650R longpass dichroic mirror and a Semrock BLP01-664R-25 longpass edge filter, respectively.
The spatial resolution of the system was measured by backilluminating a 1951 USAF resolution target with a red LED; Fig. 1(b) shows an image of group 7 of the target.For realtime visualization, pixelation artifacts due to the fiber cores were removed by convolution with a 2-D Gaussian filter (σ ¼ 1.6 pixels, 1.4 μm on the proximal face of the bundle).Figure 1(b) shows an image of the numeral "7" of the resolution target before (top) and after (bottom) removal of pixelation artifacts.Figure 1(c) demonstrates the axial sectioning strength of LS-CLE for detector slit-width of 4 μm (measured as described in Ref. 20).For this study, a slit width of 4 μm was used to give an axial resolution of ∼11 μm.While point-scanning pCLE would probably be needed for intravenous fluorescein imaging (see Ref. 20 for a discussion), our partially sectioning system is sufficient for imaging tissue sites topically stained with methylene blue and provides the advantage of a much higher frame rate.The high frame rate enables image mosaics to be assembled more readily, 21 which, as discussed in Ref. 10, is important for allowing features to be identified with confidence.
A total of 15 freshly excised tissue samples were acquired from five patients, who underwent breast surgery between March and August, 2016, under an Imperial College tissue bank license (Project R12047).Several small cut-outs, approximately 10 × 10 mm in size, were taken from normal and diseased sections (as subsequently confirmed by histology) and imaged using the following protocol ∼30 to 45 min after excision.
Methylene blue could be applied topically for in vivo imaging, as demonstrated in Ref. 19.For this ex-vivo study, methylene blue solution (0.1% in saline; Sigma Aldrich, United Kingdom) was applied topically to each cut-out for 30 s followed by a gentle wash with PBS saline to remove excess stain.The tip of the endomicroscope probe was gently applied to the tissue surface.At the end of the imaging procedure, excess methylene-blue dye was gently wiped off the surface of the tissue, and the specimen was returned to histology for routine analysis.
In postprocessing, a mosaicking algorithm was used to stitch overlapping frames, producing a larger image more comparable to histology slides.This algorithm, described fully in Ref. 21  and based on an established procedure, 22 uses two-way fast normalized cross-correlation (NCC) for pairwise image registration and combines images using distance-weighted alpha-blending.Mosaics were then stored digitally in a prospectively maintained database.To assist with probe scanning during image acquisition, a real-time preview of the mosaics, using one-way NCC and dead-leaf blending, was displayed to the operator.
The accuracy of normal/tumor identification using acriflavine staining pCLE has already been established; 10 so, the main purpose of this pilot study was to determine whether the 660-nm system could identify features following the taxonomy of pathologies developed in the previous study.A total of 450 pCLE image mosaics (mean 30 per sample; range 20 to 40) were generated from 15 specimens.Since we have a limited number of tissue samples, we acquired many more mosaics than necessary for detailed performance assessment.Figure 2 provides representative LS-CLE image mosaics of freshly excised normal, benign, and neoplastic human breast specimens stained with methylene blue.The potential of the approach to identify the microscopic tissue morphology relevant to tumor identification is demonstrated by comparison of the reconstructed mosaics with those from the previously generated data bank generated using the Cellvizio pCLE with acriflavine (0.02%) staining, and with conventional histology. 10wo features typifying normal breast tissue were identified: adipose tissue and the presence of collagen fibres in the stroma.As shown in Figs.2(a) and 2(b), the fluorescence images obtained from methylene-blue and acriflavine-stained adipose tissue are similar: the well-defined and uniform nonfluorescent polygon shape of the adipose tissue with hyperfluorescent borders can clearly be observed in both cases.Nuclei, which may not always be clearly visible, appear as sparsely populated bright spots on the borders.In Figs.2(d) and 2(e), the fibrous connective tissue appears as hyperfluorescent bundles of elastic wavy fibres.Figures 2(g) and 2(h) represent fibrocystic changes, where compressed bright slit-like glandular spaces are surrounded by loose fibrous stroma.In Figs.2(j) and 2(k), stroma composed of hyperfluoroscent spindle-shaped fibroblasts appear surrounded by gray collagen fibers.Also, both the systems enable visualization of distinctive neoplastic changes characterized by increased cellularity in the stroma, as depicted in Figs.2(m) and 2(n).The disorganized distribution of cell nuclei and the unclear tissue architecture was clearly visible as large clusters of hyperfluorescent spots.Similar features are observed in matched histology images of acriflavine-stained tissue (c, f, i, l, and o).Furthermore, we compared the signal-to-noise ratio (SNR) of the two sets of images, obtained by the ratio of the mean intensity of a homogenous stained area of 10 × 10 pixels of the tissue (signal) to the standard deviation of an unstained region of 10 × 10 pixels (noise).Averaged across 100 images, the SNR was (179 AE 10) for the acriflavine-stained images and (170 AE 13) for the methylene-blue stained images.Therefore, despite the differences in stain, imaging system and processing, the noise in the final processed images are broadly comparable.
In summary, a high-speed 660-nm LS-CLE system has been used with topical application of 0.1% methylene blue to demonstrate rapid morphological assessment of ex-vivo breast cancer tissues.Images and mosaics of normal and neoplastic breast tissue, acquired with a lateral resolution of 2.2 μm at 120 Hz, showed distinctive morphological features.The methylene blue aided LS-CLE produced images with comparable contrast to point-scanning pCLE with acriflavine staining but with the advantage of a higher frame rate and use of a safe and regulatory-approved (off-label) stain.
These current results suggest a clinical application for methylene-blue aided LS-CLE in breast cavity scanning for rapid decision-making during BCS, allowing oncological status of resection margins to be determined intraoperatively.This could allow the surgeon to accomplish more accurate and complete surgical resections, thus reducing the reoperation rates and improving outcomes.Further work will be required to confirm that the quantitative accuracy when using methylene-blue staining is similar to that when using acriflavine and to establish whether similar images can be obtained in-vivo.

Disclosures
Ms. Vyas, Dr. Hughes, Mr. Leff, and Prof. Yang have nothing to disclose.No conflicts of interest, financial or otherwise, are declared by the authors.

Fig. 1
Fig. 1 (a) Schematic layout of high-speed LS-CLE system operating at 660 nm.(b) Image of high-resolution USAF resolution target showing group 7 elements 1-6 when back-illuminated with a red LED.(c) Axial sectioning profile for the line-scanning system.