Mainly due to high spatial or temporal resolution, optical imaging techniques have been attractive tools in biomedicine with widespread research and clinical applications when compared with nonoptical methods.1 However, mainly due to the strong scattering in most nontransparent tissues, the strength of the probe light beam attenuates gradually as it penetrates deeper into the tissue. The attenuation effect fundamentally limits the available imaging performances in penetration depth, image contrast, and image resolution.23.–4 In particular, cutaneous blood perfusion is highly correlated to a number of peripheral vascular diseases, but the optical imaging of cutaneous blood vessels is quite challenging due to the high scattering property of skin. By matching the refractive index between different tissue components, tissue optical clearing has been found useful for reducing light scattering and improving imaging performance in a wide range of optical techniques, such as laser speckle contrast imaging (LSCI),5,6 optical coherence tomography (OCT),7,8 photoacoustic microscopy,9,10 confocal microscopy,11,12 two-photon microscopy,13,14 and flow cytometry.15
Extensive efforts have been devoted to the screening of high-efficiency and biocompatible optical clearing agents (OCAs) or chemical penetration enhancers for improving the diffusion of OCAs within tissues.3,4 In a previous study, we found a mixture of fructose with PEG-400 and thiazone (FPT) allows in vivo blood flow imaging with higher contrast and resolution.16 Currently, evaluation of a newly developed OCA is usually performed by imaging the blood flow in vivo5,16 or by measuring the optical properties via integration sphere,17 diffuse reflectance spectroscopy,18 and OCT7,1920.–21 separately. LSCI allows a full-field imaging of the cutaneous blood flow using speckle dynamics and has been used for evaluating the OCA-induced improvement in both the contrast and resolution of blood flow imaging. Due to its lack of depth-resolved information, LSCI fails to measure the optical properties in vivo.2223.24.–25 Spectral domain OCT (SDOCT) is capable of high-resolution, real-time, and three-dimensional (3-D) imaging of internal structure of biological tissues in a depth-resolved manner, and the contrast of OCT structure mainly originates from the endogenous differences in optical scattering, eliminating the requirement of tissue-labeling or contrast-enhancing agents.7,19,21,26 The depth-resolved OCT signal offers an approach for measuring optical properties.7,1920.–21 Thus, the attenuation coefficient, refractive-index mismatching extent, and permeation rate of different tissue compounds can be quantified, which has played a vital role in evaluating the changes of tissue optical properties in previous studies.21 OCA processing, skin optical properties, and blood flow imaging performances are closely correlated, but their relationship has seldom been reported. An approach capable of monitoring the agent-induced changes in skin optical properties and flow imaging performance simultaneously is desirable for a comprehensive understanding of a newly developed OCA.
OCT angiography (Angio-OCT) is able to generate structural and angiographic images of skin in parallel. By mathematically analyzing the temporal dynamics of light scattering, Angio-OCT is capable of contrasting the dynamic blood flow against the static tissue bed,27184.108.40.206.–32 and enables a label-free, motion-contrast 3-D microangiography. Thus, Angio-OCT is quite suitable for OCA characterization by measuring skin optical properties and evaluating blood flow imaging simultaneously.
Angio-OCT is used in this study for FPT evaluation. The optical clearing efficacy is quantified by calculating the total attenuation coefficient, refractive-index mismatching extent, and permeation rate. FPT-induced improvement in Angio-OCT imaging performance is presented through imaging resolution, contrast, and imaging depth. Further data analysis is performed to evaluate the correlation between the relative changes in Angio-OCT imaging performance and skin optical properties in vivo. Finally, since the FPT treatment enables the visualization of deep cutaneous blood vessels, dynamic study of flow response to vasoactive drugs is demonstrated by Angio-OCT.
Materials and Methods
Angio-OCT is home-built based on a typical SDOCT configuration, as shown in Fig. 1. Briefly, the light source is a broadband superluminescent diode (SLD; Superlum, Carrigtwohill, Ireland, Broadlighters D855-HP2) with a central wavelength of 850 nm and a full-width-half-maximum bandwidth of 100 nm, theoretically offering a high axial resolution of in air. An objective lens with a focal length of 40 mm was used to focus the probing light beam on the region of interest, yielding a measured lateral-resolution of . The OCT detection unit in our system was a high-speed spectrometer, equipped with a fast line-scan CMOS camera (Basler, Ahrensburg, Germany, Sprint spL4096-140k) providing a 120-kHz line-scan rate and 2048 active pixels. The spectrometer had a designed spectral resolution of 0.062 nm, providing an imaging range of on each side of the zero-delay line in air. The system sensitivity was measured at at the depth position of 0.5 mm, with an incident optical power of 2 mW upon the sample surface.
FPT was used in this study for optical clearing. As described in our previous study,16 FPT was a mixture of saturated fructose (Sinopharm Chemical Reagent Co., Ltd.) solution (78%, w/w) with PEG-400 (Kemiou Reagent Development Corporation) and thiazone (Heming Trading Company Limited) at a volume ratio of . FPT has a refractive index of 1.4801 and a pH value of 6.0. Norepinephrine (NE), purchased from Affiliated Hospital of Huazhong University of Science and Technology (Wuhan, PRC), was used for the dynamic study of cutaneous blood flow.
Male Balb/c mice (, 8 weeks old) were obtained from Shanghai Animal Experimental Center (Shanghai, PRC) and fed under specific pathogen-free conditions. Mice were anesthetized by intraperitoneal injection of a mixture of 10% urethane and 2% -chloralose with a dosage of . The dorsal hair was shaved and the residual hair was thoroughly removed with depilatory cream. Then mice were placed on the experimental platform for OCT imaging. All procedures in the experiments were carried out with the guidelines of the Institutional Animal Care and Use Committee of Zhejiang University, PRC. To evaluate the clearing efficacy of the FPT agent in vivo, the dorsal skin of 15 prepared mice was imaged with Angio-OCT before and after topical application of OCAs. Then six of these mice were injected with of NE () after skin became transparent, and the dynamic response of blood flow to the vasoactive drug was monitored for 25 min with a time interval of 1 min.
Quantification of Imaging Performance and Skin Optical Properties
OCT provides depth-resolved images of tissue. Due to the optical scattering and absorption of tissues, the OCT probe light is attenuated as the beam penetrates deeper into the tissue. The OCT depth profile reveals the depth-dependent attenuation coefficient in the tissue. According to Lambert–Beer’s law, the OCT depth profile is related to the attenuation coefficient of tissues as21,3334 proposed a new method for local attenuation coefficient estimation, which can be expressed as
The scattering coefficient of tissue is closely dependent on the refractive-index mismatching extent between extracellular fluid () and tissue components (e.g., collagen, fibers, and cell) (). In a simple model of scattering dielectric sphere, can be approximated as35
In addition, the permeability rate () is an important parameter for the evaluation of efficiency of OCAs, which can be estimated by dividing the thickness of the region that OCAs reach by the time of molecular permeation in the monitored region21
The correlation between the relative changes in refractive-index mismatching () and Angio-OCT signal strength can be calculated as
Optical Coherence Tomography Structural and Flow Imaging of Mouse Dorsal Skin
Figure 2 shows the representative OCT structural and angiographic images of mouse dorsal skin in vivo. Figure 2(a) is a cross-sectional schematic of the layered skin. Figures 2(b1) and 2(c1) show the representative OCT structural cross-sections before and after the topical application of the FPT agent, respectively. The layered structures of skin are quite vague and difficult to distinguish in Fig. 2(b1). In contrast, the imaging contrast and depth are improved greatly in Fig. 2(c1), and the layered structures and features are clearly identified and manually depicted, including the epidermis (ED), dermis (D), hypodermis (HD), and muscles (M), according to cutaneous anatomical features and the obvious boundaries in the Angio-OCT structural cross-section. The ED-D and HD-M boundaries are visualized clearly, and the dermis exhibits a decreased OCT scattering intensity. The hypodermis appears to be an interwoven structure that is correlated to the adipose cells in the layer. The muscle has a sandwich structure. After the topical application of the FPT agent, the thickness of the epidermis and dermis decreases obviously due to agent-induced dehydration. Figures 2(b2) and 2(c2) are the corresponding OCT angiograms before and after FPT treatment, respectively. The blood vessels are isolated from the tissue bed. The big vessels mainly appear at the HD layer, as indicated by the bold arrows, and the small vessels above the HD layer come from the dermis, as indicated by the thin arrows. As shown in Fig. 2(c2), most of the dermal small vessels disappear after the FPT treatment, and the HD blood vessels present an improved visibility, most likely due to the decreased scattering attenuation in the dermis, which can be also observed in the projection view of the 3-D angiography in Figs. 2(b3) and 2(c3). The dermal vasculature can be observed before the FPT treatment in Fig. 2(b3), while the HD big vessels exhibit an improved contrast after the FPT treatment in Fig. 2(c3).
Quantifications of Skin Optical Properties
The skin surface can be identified in the OCT structural cross-sections. The outermost surface was flatted to align all the depth profiles. Then, the OCT depth profile was averaged along the lateral direction so as to reduce effects of the heterogenetic distribution of the cutaneous absorbers and scatters as much as possible. As shown in Fig. 3(a), the solid blue and dashed red curves correspond to the depth profiles before and after the FPT treatment, respectively. After the FPT treatment, OCT intensity strength exhibits a decrease of in the D and HD layers, with an increase of in the M layer. The attenuation coefficient image was calculated according to Eq. (2); thereafter, the outermost surface was flattened and the depth-resolved attenuation coefficient was calculated by averaging the attenuation coefficient image along the lateral direction, as shown in Fig. 3(b). After the FPT treatment, the attenuation coefficients were lower: for the ED, D, and HD layers, and for the M layer.
As mentioned before, the relative changes in the refractive-index mismatching extent () between tissue components and extracellular fluid could be quantified using Eq. (3). As shown in Fig. 3(c), the solid blue line represents the relative changes in () after FPT treatment, indicating an average decline of in the ED, D, and HD layers, and in the muscle layer, respectively. The dashed red line is the relative changes in Angio-OCT signal strength. Thereafter, we calculated the correlation between the relative changes in refractive-index mismatching () and Angio-OCT signal strength using Eq. (5), and the results demonstrate that the correlation between the relative changes in refractive-index mismatching () and Angio-OCT signal strength are about 90% and for the superficial (ED + D + HD) layers and the deeper (M) layer, respectively. In addition, the permeability rate is using Eq. (4).
Blood Flow Dynamic Responses
With a combination of optical clearing methods, Angio-OCT exhibits enhanced performance in imaging skin blood flow, allowing cutaneous hemodynamic analysis with high spatiotemporal resolution and contrast. The vasoactive drug-induced changes of vascular diameter and flow flux are estimated over time. The flow flux is measured based on the signal strength of OCT angiograms and normalized with the baseline.
Figure 4(a) shows the M-mode Angio-OCT image of the time-lapse blood flow dynamic responses to vasoactive NE after FPT treatment, and the dynamic changes in vessel diameter, which were fitted with red curves, are obvious to see. Figure 4(b) reports the quantifications of the dynamic responses of vessel diameter and normalized blood flow. The baseline of vessel diameter is . After NE injection, the diameter exhibits a gradual decrease of up to 32.8% at the time instant , then returns to of the baseline in 25 min. Along with the dynamic change in diameter, the flow flux exhibits a gradual increase of up to at the time instant , and then returns to of the baseline in 25 min.
Discussion and Conclusion
In this study, a newly developed OCA, FPT, is evaluated with Angio-OCT by quantifying skin optical property and blood flow imaging simultaneously. Angio-OCT offers depth-resolved information about the optical properties, enabling a further understanding of the OCA. Most likely due to the different levels of OCA-induced dehydration and refractive-index matching for each skin layer, there is an evident interface at the ED-D, D-HD, and HD-M junctions, as shown in the Angio-OCT structural cross-sections and depth profiles. As plotted in Figs. 3(b) and 3(c), the D layer presents the largest decrease in scattering coefficient and refractive-index mismatching, while the M layer corresponds to a minimal change.
Mainly due to the considerable decrease of attenuation in the superficial layers, more ballistic photons could penetrate into the deeper layers, leading to an improved imaging depth and blood flow contrast. As reported in this study, the application of skin OCAs leads to an improved imaging performance for the deeper tissues (refer to Fig. 2). The imaging performance improvement is most likely caused by the OCA-induced dehydration of skin, and the reduction of attenuation coefficient [refer to Fig. 3(b)] and refractive-index mismatching within the tissue [refer to Fig. 3(c)], as indicated by the high correlation between the relative changes in refractive-index mismatching () and Angio-OCT signal strength [refer to Fig. 3(c)]. These were quite consistent with the conclusions by Ghosn et al.,36 Zhong et al.,37 Xu et al.,38 and Wen et al.39
The permeability rate can also be measured by monitoring the OCT signal slope, the method developed in Ref. 21. The optical clearing rate is an important parameter for OCA evaluation. A fast permeability rate ensures a sufficient optical clearing efficacy within a short time, which is extremely significant for the in vivo study.
As discussed above, the FPT is an efficient agent for optical clearing, inducing a significant decrease in attenuation coefficient and refractive-index mismatching, and an improved penetration with a sufficient rate. Combined with optical clearing, Angio-OCT exhibits enhanced performance in imaging skin blood flow, allowing cutaneous hemodynamic analysis with satisfactory spatiotemporal resolution and contrast, and providing a powerful approach for studying skin microcirculation.
The authors acknowledge financial support from National Hi-Tech Research and Development Program of China (2015AA020515), National Natural Science Foundation of China (Grant Nos. 61475143, 11404285, 81171376, 91232710, 31571002, 61335003, 61327007, and 61275196), Zhejiang Provincial Natural Science Foundation of China (LY14F050007), Zhejiang Province Science and Technology Grant (2015C33108), Seed project of Wuhan National Laboratory for Optoelectronics, Science Fund for Creative Research Group (Grant No. 61421064), Fundamental Research Funds for the Central Universities (2014QNA5017), and Scientific Research Foundation for Returned Scholars, Ministry of Education of China.