To analyze the formation mechanism of the halo on low light level image intensifiers and the influencing factors on the halo size, a halo tester has been designed. Under the illumination between 10-2 lx and 10-4 lx, we use the tester to collect a 0.1922 mm hole image directly with CoolSNAPK4 charge-coupled device (CCD) in a darkroom. The practical measurement result shows that the amplification ratio is 343.4. Then we put the super second and third generation image intensifiers after the hole, and the halo sizes of the hole images on the screens are determined as 0.2388 and 0.5533 mm respectively. The results are helpful to improve the quality of the low light level image intensifiers.