We have performed pulsed neutron and pulsed laser tests of a CVD diamond detector manufactured from DIAFILM, a commercial grade of CVD diamond. The laser tests were performed at the short pulse UV laser at Bechtel Nevada in Livermore, CA. The pulsed neutrons were provided by DT capsule implosions at the OMEGA laser fusion facility in Rochester, NY. From these tests, we have determined the impulse response to be 250 ps fwhm for an applied E-field of 500 V/mm. Additionally, we have determined the sensitivity to be 2.4 mA/W at 500 V/mm and 4.0 mA/W at 100 V/mm. These values are approximately 2 to 5x times higher than those reported for natural Type IIa diamond at similar E-field and thickness (1mm). These characteristics allow us to conceive of a neutron time-of-flight current mode spectrometer based on CVD diamond. Such an instrument would sit inside the laser fusion target chamber close to target chamber center (TCC), and would record neutron spectra fast enough such that backscattered neutrons and (gamma) rays from the target chamber wall would not be a concern. The acquired neutron spectra could then be used to extract DD fuel areal density from the downscattered secondary to secondary ratio.
Areal density ((sigma) R) is a fundamental parameter that characterizes the performance of an ICF implosion. For high areal densities ((sigma) R>0.1 g/cm<SUP>2</SUP>), which will be realized in implosion experiments at NIF and LMJ, the target areal density exceeds the stopping range of charged particles and measurements with charged particle spectroscopy will be difficult. In this region, an areal density measurement method using down shifted neutron counting is a promising alternative. The probability of neutron scattering in the imploded plasma is proportional to the areal density of the plasma. The spectrum of neutrons scattered by the specific target nucleus has a characteristic low energy cut off. This enables separate, simultaneous measurements of fuel and pusher (sigma) Rs. To apply this concept in implosion experiments, the detector should have extremely large dynamic range. Sufficient signal output for low energy neutrons is also required. A lithium-glass scintillation-fiber plate (LG-SCIFI) is a promising candidate for this application. In this paper we propose a novel technique based on down shifted neutron measurements with a lithium-glass sctintillation-fiber plate. The details of instrumentation and background estimation with Monte Carlo calculation are reported.
We are currently developing a high-energy (10 - 15 MeV) neutron imaging system for use in NDE applications. Our goal is to develop an imaging system capable of detecting cubic- mm-scale voids or other structural defects in heavily- shielded low-Z materials within thick sealed objects. The system will be relatively compact (suitable for use in a small laboratory) and capable of acquiring tomographic image data sets. The design of a prototype imaging detector and multi-axis staging system will be discussed and selected results from recent imaging experiments will be presented. The development of an intense, accelerator-driven neutron source suitable for use with the imaging system will also be discussed.