In this paper, an egg-shaped microbubble is proposed and analyzed firstly, which is fabricated by the pressure-assisted arc discharge technique. By tailoring the arc parameters and the position of glass tube during the fabrication process, the thinnest wall of the fabricated microbubble could reach to the level of 873nm. Then, the fiber Fabry–Perot interference technique is used to analyze the deformation of microbubble that under different filling pressures. It is found that the endface of micro-bubble occurs compression when the inner pressure increasing from 4Kpa to 1400KPa. And the pressure sensitivity of such egg-shaped microbubble sample is14.3pm/Kpa. Results of this study could be good reference for developing new pressure sensors, etc.
We proposed and experimentally demonstrated four kinds of high-sensitivity gas pressure sensors based on in-fiber devices, including a sub-micron silica diaphragm-based fiber-tip, a polymer-capped Fabry-Perot interferometer, an inflated long period fiber grating and a twin core fiber-based Mach-Zehnder interferometer, which have sensitivities of 1036, 1130, 1680, 9600 pm/MPa, respectively.
We demonstrated an ultrasensitive temperature sensor based on a unique fiber Fabry-Perot interferometer (FPI). The FPI was created by means of splicing a mercury-filled silica tube with a single-mode fiber (SMF). The FPI had an air cavity, which was formed by the end face of the SMF and that of the mercury column. Experimental results showed that the FPI had an ultrahigh temperature-sensitivity of up to -41 nm/°C, which was about one order of magnitude higher than those of the reported FPI-based fiber tip sensors. Such a FPI temperature sensor is expected to have potential applications for highly-sensitive ambient temperature sensing.
We investigated experimentally liquid crystal (LC) filled photonic crystal fiber’s temperature responses at different temperature ranging from 30 to 80°C. Experimental evidences presented that the LC’s clearing point temperature was 58°C, which is consistent with the theoretical given value. The bandgap transmission was found to have opposite temperature responses lower and higher than the LC’s clearing point temperature owing to its phase transition property. A high bandgap tuning sensitivity of 105 nm/°C was achieved around LC’s clearing point temperature.
We demonstrated a high-sensitivity strain sensor based on an in-line Fabry-Perot interferometer with an air cavity whose was created by splicing together two sections of standard single mode fibers. The sensitivity of this strain sensor was enhanced to 6.02 pm/με by improving the cavity length of the Fabry-Perot interferometer by means of repeating arc discharges for reshaping the air cavity. Moreover, such a strain sensor has a very low temperature sensitivity of 1.06 pm/°C, which reduces the cross-sensitivity problem between tensile strain and temperature.
An improved arc discharge technique was demonstrated to inscribe high-quality LPFGs with a resonant attenuation of - 28 dB and an insertion loss of 0.2 dB by use of a commercial fusion splicer. Such a technique avoids the influence of the mass which is prerequisite for traditional technique. Moreover, no physical deformation was observed on the LPFG surface. Compared with more than 86 grating periods required by traditional arc discharge technique, only 27 grating periods were required to inscribe a compact LPFG by our improved arc discharge technique.