Photonic ring resonators used as wavelength notch filters are a promising novel solution to enable astronomical instruments to remove the signal from atmospheric OH emission in the near-infrared wavelength range. We derive design requirements from theory and finite difference time domain simulations. We find rings with radii less than 10 microns provide an adequate free spectral range for silicon nitride abd less than 3 microns for silicon. One challenge for this application is the requirement for many rings in series to suppress particular wavelengths within 0.2nm. We report progress in fabricating both silicon and silicon nitride rings for OH suppression.
Integrated optics has the potential to play a transformative role in astronomical instrumentation. It has already made a significant impact in the field of optical interferometry, through the use of planar waveguide arrays for beam combination and phase-shifting. Additionally, the potential benefits of micro-spectrographs based on array waveguide gratings have also been demonstrated.
Here we examine a new application of integrated optics, using ring resonators as notch filters to remove the signal from atmospheric OH emission lines from astronomical spectra. We also briefly discuss their use as frequency combs for wavelength calibration and as drop filters for Doppler planet searches. We discuss the theoretical requirements for ring resonators for OH suppression. We find that small radius (< 10 μm), high index contrast (Si or Si3N4) rings are necessary to provide an adequate free spectral range. The suppression depth, resolving power, and throughput for efficient OH suppression can be realised with critically coupled rings with high self-coupling coefficients.
We report on preliminary laboratory tests of our Si and Si3N4 rings and give details of their fabrication. We demonstrate high self-coupling coefficients (> 0:9) and good control over the free spectral range and wavelength separation of multi-ring devices. Current devices have Q ≈ 4000 and ≈ 10 dB suppression, which should be improved through further optimisation of the coupling coefficients. The overall prospects for the use of ring resonators in astronomical instruments is promising, provided efficient fibre-chip coupling can be achieved.
The Dark Energy Survey (DES) is an operating optical survey aimed at understanding the accelerating expansion of the universe using four complementary methods: weak gravitational lensing, galaxy cluster counts, baryon acoustic oscillations, and Type Ia supernovae. To perform the 5000 sq-degree wide field and 30 sq-degree supernova surveys, the DES Collaboration built the Dark Energy Camera (DECam), a 3 square-degree, 570-Megapixel CCD camera that was installed at the prime focus of the Blanco 4-meter telescope at the Cerro Tololo Inter-American Observatory (CTIO). DES has completed its third observing season out of a nominal five. This paper describes DES “Year 1” (Y1) to “Year 3” (Y3), the strategy, an outline of the survey operations procedures, the efficiency of operations and the causes of lost observing time. It provides details about the quality of the first three season's data, and describes how we are adjusting the survey strategy in the face of the El Niño Southern Oscillation.
The Dark Energy Survey (DES) is a next generation optical survey aimed at understanding the accelerating expansion of the universe using four complementary methods: weak gravitational lensing, galaxy cluster counts, baryon acoustic oscillations, and Type Ia supernovae. To perform the 5000 sq-degree wide field and 30 sq-degree supernova surveys, the DES Collaboration built the Dark Energy Camera (DECam), a 3 square-degree, 570-Megapixel CCD camera that was installed at the prime focus of the Blanco 4-meter telescope at the Cerro Tololo Inter-American Observatory (CTIO). DES started its first observing season on August 31, 2013 and observed for 105 nights through mid-February 2014. This paper describes DES “Year 1” (Y1), the strategy and goals for the first year's data, provides an outline of the operations procedures, lists the efficiency of survey operations and the causes of lost observing time, provides details about the quality of the first year's data, and hints at the “Year 2” plan and outlook.
The Dark Energy Survey Collaboration has completed construction of the Dark Energy Camera (DECam), a 3 square
degree, 570 Megapixel CCD camera which will be mounted on the Blanco 4-meter telescope at CTIO. DECam will be
used to perform the 5000 sq. deg. Dark Energy Survey with 30% of the telescope time over a 5 year period. During the
remainder of the time, and after the survey, DECam will be available as a community instrument. All components of
DECam have been shipped to Chile and post-shipping checkout finished in Jan. 2012. Installation is in progress. A
summary of lessons learned and an update of the performance of DECam and the status of the DECam installation and
commissioning will be presented.
The Dark Energy Survey Camera (DECam) will be comprised of a mosaic of 74 charge-coupled devices (CCDs). The
Dark Energy Survey (DES) science goals set stringent technical requirements for the CCDs. The CCDs are provided by
LBNL with valuable cold probe data at 233 K, providing an indication of which CCDs are more likely to pass. After
comprehensive testing at 173 K, about half of these qualify as science grade. Testing this large number of CCDs to
determine which best meet the DES requirements is a very time-consuming task. We have developed a multistage
testing program to automatically collect and analyze CCD test data. The test results are reviewed to select those CCDs
that best meet the technical specifications for charge transfer efficiency, linearity, full well capacity, quantum efficiency,
noise, dark current, cross talk, diffusion, and cosmetics.
The Dark Energy Camera is an wide field imager currently
under construction for the Dark Energy Survey.
This instrument will use fully depleted 250 μm thick
CCD detectors selected for their higher quantum efficiency
in the near infrared with respect to thinner devices.
The detectors were developed by LBNL using
high resistivity Si substrate. The full set of scientific
detectors needed for DECam has now been fabricated,
packaged and tested. We present here the results of
the testing and characterization for these devices and
compare these results with the technical requirements
for the Dark Energy Survey.
The Dark Energy Survey Collaboration is building the Dark Energy Camera (DECam), a 3 square degree, 520
Megapixel CCD camera which will be mounted on the Blanco 4-meter telescope at CTIO. DECam will be used to
perform the 5000 sq. deg. Dark Energy Survey with 30% of the telescope time over a 5 year period. During the
remainder of the time, and after the survey, DECam will be available as a community instrument. Construction of
DECam is well underway. Integration and testing of the major system components has already begun at Fermilab and
the collaborating institutions.
DECam is a 520 Mpix, 3 square-deg FOV imager being built for the Blanco 4m Telescope at CTIO. This facility
instrument will be used for the "Dark Energy Survey" of the southern galactic cap. DECam has chosen 250 μm thick
CCDs, developed at LBNL, with good QE in the near IR for the focal plane. In this work we present the characterization
of these detectors done by the DES team, and compare it to the DECam technical requirements. The results demonstrate
that the detectors satisfy the needs for instrument.
We describe the Dark Energy Camera (DECam), which will be the primary instrument used in the Dark Energy Survey.
DECam will be a 3 sq. deg. mosaic camera mounted at the prime focus of the Blanco 4m telescope at the Cerro-Tololo
International Observatory (CTIO). DECam includes a large mosaic CCD focal plane, a five element optical corrector,
five filters (g,r,i,z,Y), and the associated infrastructure for operation in the prime focus cage. The focal plane consists of
62 2K x 4K CCD modules (0.27"/pixel) arranged in a hexagon inscribed within the roughly 2.2 degree diameter field of
view. The CCDs will be 250 micron thick fully-depleted CCDs that have been developed at the Lawrence Berkeley
National Laboratory (LBNL). Production of the CCDs and fabrication of the optics, mechanical structure, mechanisms,
and control system for DECam are underway; delivery of the instrument to CTIO is scheduled for 2010.