The Greenland Telescope project has recently participated in an experiment to image the supermassive black hole shadow at the center of M87 using Very Long Baseline Interferometry technique in April of 2018. The antenna consists of the 12-m ALMA North American prototype antenna that was modified to support two auxiliary side containers and to withstand an extremely cold environment. The telescope is currently at Thule Air Base in Greenland with the long-term goal to move the telescope over the Greenland ice sheet to Summit Station. The GLT currently has a single cryostat which houses three dual polarization receivers that cover 84-96 GHz, 213-243 GHz and 271-377 GHz bands. A hydrogen maser frequency source in conjunction with high frequency synthesizers are used to generate the local oscillator references for the receivers. The intermediate frequency outputs of each receiver cover 4-8 GHz and are heterodyned to baseband for digitization within a set of ROACH-2 units then formatted for recording onto Mark-6 data recorders. A separate set of ROACH-2 units operating in parallel provides the function of auto-correlation for real-time spectral analysis. Due to the stringent instrumental stability requirements for interferometry a diagnostic test system was incorporated into the design. Tying all of the above equipment together is the fiber optic system designed to operate in a low temperature environment and scalable to accommodate a larger distance between the control module and telescope for Summit Station. A report on the progress of the above electronics instrumentation system will be provided.
The Greenland Telescope (GLT) project and the East Asian Observatory (EAO) successfully commissioned the first light GLT instrument at the James Clerk Maxwell Telescope (JCMT) in Hawaii, prior to transferring the instrument to Greenland. The GLT instrument which comprises of a cryostat with three cartridge-type receivers (at 86GHz, 230GHz and 345GHz) was installed into the receiver cabin of JCMT and operated in three modes: - (a) Regular JCMT observing with the GLT instrument, using ACSIS, (JCMT’s autocorrelation spectrometer) as the backend and JCMT software for telescope control, data reduction, pointing and antenna focus adjustment. (b) Single dish observations of astronomical spectral line sources, recording data onto mark 6 recorders for offline data reduction. (c) eSMA interferometer array observations at 230GHz in conjunction with the SMA. In this paper, we report on the installation and integration of the GLT instrument at JCMT, present results from commissioning and show how the success of the GLT instrument commissioning fits with our plans for future instrumentation at JCMT.
The Greenland Telescope completed its construction, so the commissioning phase has been started since December 2017. Single-dish commissioning has started from the optical pointing which produced the first pointing model, followed by the radio pointing and focusing using the Moon for both the 86 GHz and the 230 GHz receivers. After Venus started to rise from the horizon, the focus positions has been improved for both receivers. Once we started the line pointing using the SiO(2-1) maser line and the CO(2-1) line for the 86 GHz and the 230 GHz receivers, respectively, the pointing accuracy also improved, and the final pointing accuracy turned to be around 3" - 5" for both receivers. In parallel, VLBI commissioning has been performed, with checking the frequency accuracy and the phase stability for all the components that would be used for the VLBI observations. After all the checks, we successfully joined the dress rehearsals and actual observations of the 86 GHz and 230 GHz VLBI observations, The first dress rehearsal data between GLT and ALMA were correlated, and successfully detected the first fringe, which confirmed that the GLT commissioning was successfully performed.
The Greenland Telescope Project (GLT) has successfully commissioned its 12-m sub-millimeter. In January 2018, the fringes were detected between the GLT and the Atacama Large Millimeter Array (ALMA) during a very-long-baseline interferometry (VLBI) exercise. In April 2018, the telescope participated in global VLBI science observations at Thule Air Base (TAB). The telescope has been completely rebuilt, with many new components, from the ALMA NA (North America) Prototype antenna and equipped with a new set of sub-millimeter receivers operating at 86, 230, and 345 GHz, as well as a complete set of instruments and VLBI backends. This paper describes our progress and status of the project and its plan for the coming decade.
We describe the control and monitoring system for the Greenland Telescope (GLT). The GLT is a 12-m radio telescope aiming to carry out the sub-millimeter Very Long Baseline Interferometry (VLBI) observations and image the shadow of the super massive black hole in M87. In November 2017 construction has been finished and commissioning activity has been started. In April 2018 we participated in the VLBI observing campaign for the Event Horizon Telescope (EHT) collaboration. In this paper we present the entire GLT control/monitoring system in terms of computers, network and software.
The Greenland Telescope (GLT) is a 12m diameter antenna that is being developed from the ALMA North America prototype antenna, for VLBI observations and single-dish science approaching THz, at the Summit station in Greenland. The GLT is a collaboration between the Smithsonian Astrophysical Observatory and the Academia Sinica Institute of Astronomy and Astrophysics. We describe the control and monitoring software that is being developed for GLT. The present version of the software is ready for the initial tests of the antenna at Thule, including optical and radio pointing calibration, holography, and VLBI observations at 230 GHz.
Since the ALMA North America Prototype Antenna was awarded to the Smithsonian Astrophysical Observatory (SAO), SAO and the Academia Sinica Institute of Astronomy and Astrophysics (ASIAA) are working jointly to relocate the antenna to Greenland. This paper shows the status of the antenna retrofit and the work carried out after the recommissioning and subsequent disassembly of the antenna at the VLA has taken place. The next coming months will see the start of the antenna reassembly at Thule Air Base. These activities are expected to last until the fall of 2017 when commissioning should take place. In parallel, design, fabrication and testing of the last components are taking place in Taiwan.
The Greenland Telescope project will deploy and operate a 12m sub-millimeter telescope at the highest point of the Greenland i e sheet. The Greenland Telescope project is a joint venture between the Smithsonian As- trophysical Observatory (SAO) and the Academia Sinica Institute of Astronomy and Astrophysics (ASIAA). In this paper we discuss the concepts, specifications, and science goals of the instruments being developed for single-dish observations with the Greenland Telescope, and the coupling optics required to couple both them and the mm-VLBI receivers to antenna. The project will outfit the ALMA North America prototype antenna for Arctic operations and deploy it to Summit Station,1 a NSF operated Arctic station at 3,100m above MSL on the Greenland I e Sheet. This site is exceptionally dry, and promises to be an excellent site for sub-millimeter astronomical observations. The main science goal of the Greenland Telescope is to carry out millimeter VLBI observations alongside other telescopes in Europe and the Americas, with the aim of resolving the event horizon of the super-massive black hole at the enter of M87. The Greenland Telescope will also be outfitted for single-dish observations from the millimeter-wave to Tera-hertz bands. In this paper we will discuss the proposed instruments that are currently in development for the Greenland Telescope - 350 GHz and 650 GHz heterodyne array receivers; 1.4 THz HEB array receivers and a W-band bolometric spectrometer. SAO is leading the development of two heterodyne array instruments for the Greenland Telescope, a 48- pixel, 325-375 GHz SIS array receiver, and a 4 pixel, 1.4 THz HEB array receiver. A key science goal for these instruments is the mapping of ortho and para H2D+ in old protostellar ores, as well as general mapping of CO and other transitions in molecular louds. An 8-pixel prototype module for the 350 GHz array is currently being built for laboratory and operational testing on the Greenland Telescope. Arizona State University are developing a 650 GHz 256 pixel SIS array receiver based on the KAPPa SIS mixer array technology and ASIAA are developing 1.4 THz HEB single pixel and array receivers. The University of Cambridge and SAO are collaborating on the development of the CAMbridge Emission Line Surveyor (CAMELS), a W-band `on- hip' spectrometer instrument with a spectral resolution of R ~ 3000. CAMELS will consist of two pairs of horn antennas, feeding super conducting niobium nitride filter banks read by tantalum based Kinetic Inductance Detectors.
The ALMA North America Prototype Antenna was awarded to the Smithsonian Astrophysical Observatory (SAO) in 2011. SAO and the Academia Sinica Institute of Astronomy and Astrophysics (ASIAA), SAO’s main partner for this project, are working jointly to relocate the antenna to Greenland to carry out millimeter and submillimeter VLBI observations. This paper presents the work carried out on upgrading the antenna to enable operation in the Arctic climate by the GLT Team to make this challenging project possible, with an emphasis on the unexpected telescope components that had to be either redesigned or changed. Five-years of inactivity, with the antenna laying idle in the desert of New Mexico, coupled with the extreme weather conditions of the selected site in Greenland have it necessary to significantly refurbish the antenna. We found that many components did need to be replaced, such as the antenna support cone, the azimuth bearing, the carbon fiber quadrupod, the hexapod, the HVAC, the tiltmeters, the antenna electronic enclosures housing servo and other drive components, and the cables. We selected Vertex, the original antenna manufacturer, for the main design work, which is in progress. The next coming months will see the major antenna components and subsystems shipped to a site of the US East Coast for test-fitting the major antenna components, which have been retrofitted. The following step will be to ship the components to Greenland to carry out VLBI
The Array for Microwave Background Anisotropy (AMiBA) is a radio interferometer for research in cosmology,
currently operating 7 0.6m diameter antennas co-mounted on a 6m diameter platform driven by a hexapod
mount. AMiBA is currently the largest hexapod telescope. We briefly summarize the hexapod operation with
the current pointing error model. We then focus on the upcoming
13-element expansion with its potential
difficulties and solutions. Photogrammetry measurements of the platform reveal deformations at a level which
can affect the optical pointing and the receiver radio phase. In order to prepare for the 13-element upgrade, two
optical telescopes are installed on the platform to correlate optical pointing tests. Being mounted on different
locations, the residuals of the two sets of pointing errors show a characteristic phase and amplitude difference
as a function of the platform deformation pattern. These results depend on the telescope's azimuth, elevation
and polarization position. An analytical model for the deformation is derived in order to separate the local
deformation induced error from the real hexapod pointing error. Similarly, we demonstrate that the deformation
induced radio phase error can be reliably modeled and calibrated, which allows us to recover the ideal synthesized
beam in amplitude and shape of up to 90% or more. The resulting array efficiency and its limits are discussed
based on the derived errors.
Using the array of seven 0.6m antennas in Hawaii, we have conducted short observations on several galaxy clusters through
the Sunyaev-Zeldovich effect at 3mm wavelength in 2007. The observations were done with a resolution of 6', and we
have chosen the low redshift (z=0.09-0.32) massive clusters to optimize detection. Major contamination to the data comes
from instrumental offset and ground pickup. We will demonstrate the results based on a simple on source - off source
switching observing scheme. In addition, the performance of a wideband analog 4-lag correlator was also investigated.
The Academia Sinica, Institute for Astronomy and Astrophysics (ASIAA) is installing the AMiBA interferometric array telescope at the Mauna Loa Observatory, Hawaii. The 6-meter carbon fiber fully steerable platform is mounted on the Hexapod Mount. After integration and equipment with dummy weights, the platform has been measured by photogrammetry to verify its behavior predicted by Finite Element Analysis. The Hexapod servo control is now operational and equipment of the platform with the initial 7 60-cm dishes, the correlator and electronics is underway. Pointing has started with the aid of the optical telescope. We present the status of the telescope after the servo and initial pointing tests have been carried out. We also present the results of platform measurements by photogrammetry.
AMiBA, as a dual-polarization 86-102 GHz interferometer array, is designed to measure the power spectrum of fluctuations in the cosmic microwave background (CMB) radiation, and to detect the high-redshift clusters of galaxies via the Sunyaev-Zel'dovich Effect (SZE). The operation of AMiBA is about to begin after installation of the first two receivers and correlators onto the 6-meter diameter platform by the end of 2005. The initial setup of the array will consist of 7 antennas with 60 cm diameter reflectors in a hexagonal configuration, aiming at multipoles l ~ 3000. Signals from receivers are cross-correlated in analog lag correlators. The initial operation will focus on characterizing the systematics by observing various known objects on the sky. The expansion to 13 elements with larger dishes will commence once the 7-element array testing is completed.