A Graphene Oxide (GO) modified surface plasmon resonance (SPR) sensor based on the silver-coated side-polished fiber was demonstrated. Stable GO aqueous dispersion was prepared through sonication, confirmed by UV-vis absorption spectrum and Tyndall effect. GO nanosheets were decorated on the Octadecanethiol (ODT)-silver sensor surface, where ODT act as a link between the silver and GO nano-films. The GO decoration process was in-situ monitored by SPR wavelength interrogation method. The proposed SPR fiber sensor show a refractive index sensitivity up to 2252.0 nm/RIU in the range of 1.30 ∼ 1.40 RIU and can be used as promising candidate in biosensing.
This paper reports a new silica fiber-tip Fabry-Perot interferometer with thin film and large surface area characteristic for high pressure and vacuum degree detection simultaneously, which is fabricated by etching a flat fiber tip into concave surface firstly, with subsequent arc jointing the concave fiber into a inline Fabry-Perot cavity, then drawing one surface of the F-P cavity into several micrometers scale by arc discharge and finally etching the surface into sub-micrometer scale integrally. As the silica fiber-tip Fabry-Perot interferometer film thickness could be tailored very thinly by HF acid solution, plus the surface area of thin film could be expanded during the chemical etching process, the variation of the bubble cavity length is very sensitive to the inner/outer pressure difference of the fiber-tip Fabry-Perot interferometer. Experimental result shows an high sensitivity of 780nm/MPa is feasible. Such configuration has the advantages of lowcost, ease of fabrication and compact size, which make it a promising candidate for pressure and vacuum measurement.
We investigated experimentally liquid crystal (LC) filled photonic crystal fiber’s temperature responses at different temperature ranging from 30 to 80°C. Experimental evidences presented that the LC’s clearing point temperature was 58°C, which is consistent with the theoretical given value. The bandgap transmission was found to have opposite temperature responses lower and higher than the LC’s clearing point temperature owing to its phase transition property. A high bandgap tuning sensitivity of 105 nm/°C was achieved around LC’s clearing point temperature.
We demonstrated a high-sensitivity strain sensor based on an in-line Fabry-Perot interferometer with an air cavity whose was created by splicing together two sections of standard single mode fibers. The sensitivity of this strain sensor was enhanced to 6.02 pm/με by improving the cavity length of the Fabry-Perot interferometer by means of repeating arc discharges for reshaping the air cavity. Moreover, such a strain sensor has a very low temperature sensitivity of 1.06 pm/°C, which reduces the cross-sensitivity problem between tensile strain and temperature.