The Greenland Telescope project has recently participated in an experiment to image the supermassive black hole shadow at the center of M87 using Very Long Baseline Interferometry technique in April of 2018. The antenna consists of the 12-m ALMA North American prototype antenna that was modified to support two auxiliary side containers and to withstand an extremely cold environment. The telescope is currently at Thule Air Base in Greenland with the long-term goal to move the telescope over the Greenland ice sheet to Summit Station. The GLT currently has a single cryostat which houses three dual polarization receivers that cover 84-96 GHz, 213-243 GHz and 271-377 GHz bands. A hydrogen maser frequency source in conjunction with high frequency synthesizers are used to generate the local oscillator references for the receivers. The intermediate frequency outputs of each receiver cover 4-8 GHz and are heterodyned to baseband for digitization within a set of ROACH-2 units then formatted for recording onto Mark-6 data recorders. A separate set of ROACH-2 units operating in parallel provides the function of auto-correlation for real-time spectral analysis. Due to the stringent instrumental stability requirements for interferometry a diagnostic test system was incorporated into the design. Tying all of the above equipment together is the fiber optic system designed to operate in a low temperature environment and scalable to accommodate a larger distance between the control module and telescope for Summit Station. A report on the progress of the above electronics instrumentation system will be provided.
The Greenland Telescope (GLT) project and the East Asian Observatory (EAO) successfully commissioned the first light GLT instrument at the James Clerk Maxwell Telescope (JCMT) in Hawaii, prior to transferring the instrument to Greenland. The GLT instrument which comprises of a cryostat with three cartridge-type receivers (at 86GHz, 230GHz and 345GHz) was installed into the receiver cabin of JCMT and operated in three modes: - (a) Regular JCMT observing with the GLT instrument, using ACSIS, (JCMT’s autocorrelation spectrometer) as the backend and JCMT software for telescope control, data reduction, pointing and antenna focus adjustment. (b) Single dish observations of astronomical spectral line sources, recording data onto mark 6 recorders for offline data reduction. (c) eSMA interferometer array observations at 230GHz in conjunction with the SMA. In this paper, we report on the installation and integration of the GLT instrument at JCMT, present results from commissioning and show how the success of the GLT instrument commissioning fits with our plans for future instrumentation at JCMT.
We describe the control and monitoring system for the Greenland Telescope (GLT). The GLT is a 12-m radio telescope aiming to carry out the sub-millimeter Very Long Baseline Interferometry (VLBI) observations and image the shadow of the super massive black hole in M87. In November 2017 construction has been finished and commissioning activity has been started. In April 2018 we participated in the VLBI observing campaign for the Event Horizon Telescope (EHT) collaboration. In this paper we present the entire GLT control/monitoring system in terms of computers, network and software.
The Greenland Telescope completed its construction, so the commissioning phase has been started since December 2017. Single-dish commissioning has started from the optical pointing which produced the first pointing model, followed by the radio pointing and focusing using the Moon for both the 86 GHz and the 230 GHz receivers. After Venus started to rise from the horizon, the focus positions has been improved for both receivers. Once we started the line pointing using the SiO(2-1) maser line and the CO(2-1) line for the 86 GHz and the 230 GHz receivers, respectively, the pointing accuracy also improved, and the final pointing accuracy turned to be around 3" - 5" for both receivers. In parallel, VLBI commissioning has been performed, with checking the frequency accuracy and the phase stability for all the components that would be used for the VLBI observations. After all the checks, we successfully joined the dress rehearsals and actual observations of the 86 GHz and 230 GHz VLBI observations, The first dress rehearsal data between GLT and ALMA were correlated, and successfully detected the first fringe, which confirmed that the GLT commissioning was successfully performed.
The Greenland Telescope Project (GLT) has successfully commissioned its 12-m sub-millimeter. In January 2018, the fringes were detected between the GLT and the Atacama Large Millimeter Array (ALMA) during a very-long-baseline interferometry (VLBI) exercise. In April 2018, the telescope participated in global VLBI science observations at Thule Air Base (TAB). The telescope has been completely rebuilt, with many new components, from the ALMA NA (North America) Prototype antenna and equipped with a new set of sub-millimeter receivers operating at 86, 230, and 345 GHz, as well as a complete set of instruments and VLBI backends. This paper describes our progress and status of the project and its plan for the coming decade.