The Focusing Optics X-ray Solar Imager (FOXSI) sounding rocket experiment aims to investigate fundamental questions about the high-energy Sun through direct imaging and spectroscopy of hard X-rays. The experiment utilizes Wolter-I type nested hard X-ray mirrors and fine-pitch semiconductor detectors, which are separated by a 2m focal length. Tol date, FOXSI has had two successful flights, on 2012 November 02 and 2014 December 11, demonstrating that the technology can measure small-scale energy releases (microflares and aggregated nanoflares) from the solar corona. The third flight for FOXSI is scheduled for August 2018. Significant improvements have been made on the FOXSI instrumentation, including upgraded optic modules with more nested mirror shells; specially designed collimators to mitigate the number of single bounce photons (ie., ghost rays) reaching the focal plane detector; and fine-pitch double-sided CdTe strip detectors to replace some of the Si-based hard X-ray detectors for better efficiency for hard X-rays. Furthermore, a CMOS based soft X-ray (SXR) instrument, “Phoenix”, will be added to FOXSI-3 by replacing one hard X-ray detector with a photon-counting SXR sensor. This will enable evaluation of the Sun via imaging spectroscopy simultaneously over a large X-ray energy range covering soft to hard X-rays. This paper will describe the overall instrument design of the FOXSI-3 experiment, which will be sensitive to solar soft and hard X-rays in the 1 – 20 keV range, as well as give a summary of insightful results and lessons from the first two flights. Possible observations for FOXSI-3 will also be discussed.
A simple design of a soft x-ray polarimeter using multilayer mirrors is presented. A multilayer mirror acts as a linear polarization analyzer for x-rays at incidence angles close to the Brewster angle. The instrument consists of an x-ray concentrator, a set of multilayer mirrors placed at 45 deg from the optical axis, and a detector at Nasmyth focus. The instrument rotating about its optical axis during observations can measure the linear polarization of 0.2- to 0.7-keV x-rays from astronomical sources. The use of a soft x-ray concentrator with geometrical area ∼630 cm2 provides sufficient sensitivity to address key scientific questions. Five different multilayer mirrors placed on a rotating wheel provide the option to measure polarization in any of the five narrow bands spanning the 0.2- to 0.7-keV range. Design and estimated performance of the design are discussed.
The Focusing Optics X-ray Solar Imager (FOXSI) sounding rocket experiment conducts direct imaging and spectral observation of the Sun in hard X-rays, in the energy range 4 to 20 keV. These high-sensitivity observations are used to study particle acceleration and coronal heating. FOXSI is designed with seven grazing incidence optics modules that focus X-rays onto seven focal plane detectors kept at a 2m distance. FOXSI-1 was flown with seven Double-sided Si Strip Detectors (DSSD), and two of them were replaced with CdTe detectors for FOXSI-2. The upcoming FOXSI-3 flight will carry DSSD and CdTe detectors with upgraded optics for enhanced sensitivity. The detectors are calibrated using various radioactive sources. The detector’s spectral response matrix was constructed with diagonal elements using a Gaussian approximation with a spread (sigma) that accounts for the energy resolution of the detector. Spectroscopic studies of past FOXSI flight data suggest that the inclusion of lower energy X-rays could better constrain the spectral modeling to yield a more precise temperature estimation of the hot plasma. This motivates us to carry out an improved calibration to better understand the finer-order effects on the spectral response, especially at lower energies. Here we report our improved calibration of FOXSI detectors using experiments and Monte-Carlo simulations.
In high energy solar astrophysics, imaging hard X-rays by direct focusing offers higher dynamic range and greater sensitivity compared to past techniques that used indirect imaging. The Focusing Optics X-ray Solar Imager (FOXSI) is a sounding rocket payload that uses seven sets of nested Wolter-I figured mirrors together with seven high-sensitivity semiconductor detectors to observe the Sun in hard X-rays through direct focusing. The FOXSI rocket has successfully flown twice and is funded to fly a third time in summer 2018. The Wolter-I geometry consists of two consecutive mirrors, one paraboloid and one hyperboloid, that reflect photons at grazing angles. Correctly focused X-rays reflect once per mirror segment. For extended sources, like the Sun, off-axis photons at certain incident angles can reflect on only one mirror and still reach the focal plane, generating a background pattern of singly reflected rays (i.e., ghost rays) that can limit the sensitivity of the observation to faint, focused sources. Understanding and mitigating the impact of the singly reflected rays on the FOXSI optical modules will maximize the instruments’ sensitivity to background-limited sources. We present an analysis of the FOXSI singly reflected rays based on ray-tracing simulations and laboratory measurements, as well as the effectiveness of different physical strategies to reduce them.
We present the formulation of an analytical model which simulates charge transport in Swept Charge Devices
(SCDs) to understand the nature of the spectral redistribution function (SRF). We attempt to construct the
energy-dependent and position dependent SRF by modeling the photon interaction, charge cloud generation and
various loss mechanisms viz., recombination, partial charge collection and split events. The model will help in
optimizing event selection, maximize event recovery and improve spectral modeling for Chandrayaan-2 (slated
for launch in 2014). A proto-type physical model is developed and the algorithm along with its results are
discussed in this paper.