Image Plates as x-ray recorder have advantages including reusable, high dynamic range, large active area, and so on. In this work, Fuji BAS-SR type image plate combined with BAS-5000 scanner is calibrated. The fade rates of Image Plates has been measured using x-ray diffractometric in different room temperature; the spectral response of Image Plates has been measured using <sup>241</sup>Am radioactive sealed source and fitting with linear model; the spatial resolution of Image Plates has been measured using micro-focus x-ray tube. The results show that Image Plates has an exponent decade curve and double absorption edge response curve. The spatial resolution of Image Plates with 25μ/50μ scanner resolution is 6.5lp/mm, 11.9lp/mm respectively and gold grid radiography is collected with 80lp/mm spatial resolution using SR-type Image Plates. BAS-SR type Image Plates can do high spatial resolution and quantitative radiographic works. It can be widely used in High energy density physics (HEDP), inertial confinement fusion (ICF) and laboratory astronomy physics.
Areal density (ρR) is one of the crucial parameters in the inertial confinement fusion. Measurement of the fusion products is a more feasible method to diagnose ρR than other methods, such as X-ray. In the capsules filled with D<sub>2</sub> fuel or D-<sup>3</sup>He fuel, proton is an ideal probe to diagnose the implosion ρR in different emission times and directions by measurements of the proton yields and spectra. By D-D reaction protons and D-<sup>3</sup>He reaction protons, the diagnostics of the total and fuel ρR, ρR evolution, implosion asymmetry and mix effect have been demonstrated at OMEGA and NIF facilities. Also some advanced proton diagnostics instruments are developed with a high level capability. Preliminary diagnosis of ρR in the deuterium involved fuel capsules by measurement of protons at SG-III facility was implemented. A fusion product emission and transport code by Monte-Carlo method was developed. The primary and secondary protons emission and transport in the fuel and shell plasmas were able to be simulated. The relations of the proton energy loss and the secondary proton yields with the areal density were inspected. Several proton spectrometers have been built up at SG-III facility, such as a step ranged filter (SRF) proton spectrometer and a wedged range filter (WRF) proton spectrometer. Some proton response simulation codes and the codes for proton spectra reconstruction were also developed. The demonstrations of ρR diagnostics at SG-III facility by D-D reaction and D-<sup>3</sup>He reaction proton spectra measurements are presented.