Temperature is an important factor affecting the performance of TO package LD. In order to ensure the safe and stable operation of LD, a temperature control circuit for LD based on PID technology is designed. The MAX1978 and an external PID circuit are used to form a control circuit that drives the thermoelectric cooler (TEC) to achieve control of temperature and the external load can be changed. The system circuit has low power consumption, high integration and high precision,and the circuit can achieve precise control of the LD temperature. Experiment results show that the circuit can achieve effective and stable control of the laser temperature.
The inhomogeneity of scattering medium distorts the propagation of the waves, which has been detrimental to the performance of optical imaging. The operating time of the traditional solutions will be very long as the scanning is necessary during the imaging. A recovery solution based on spatial optical transmission matrix has been proposed. With the acquiring of the spatial optical transmission matrix, the incident object wave will be recovered directly from the distorted transmitted wave, in this way, only a single shot is needed during the imaging. The effectiveness of this method has been proved by the simulation and experiment, the principle is simpler and the algorithm is more efficient, which are beneficial to the imaging through the scattering medium.
Thermal therapy (or hyperthermia) is one of the effective operations for tumor treating and curing. As tumor tissues are more susceptible to heat than normal tissues, in thermal therapy operations, temperature on operation area is a crucial parameter for optimal treating. When the temperature is too low, the tumor tissues cannot be killed; otherwise, the temperature is too high, the operation may damage normal tissues around the tumor. During thermal therapy operation, the heating power is normally supplied by high-frequency EM field, so traditional temperature sensors, such as thermal couples, thermistors, cannot work stably due to EM interference. We present a multi-function endoscope optical fiber temperature sensor system. With this sensor setup based on principle of fluorescence life time, the temperature on operation point is detected in real time. Furthermore, a build-in endoscope centers in the fiber sensor, thus the operation area can be viewed or imaged directly during the operation. This design can navigate the operation, particularly for in vivo operations. The temperature range of the sensor system is 30°C-150°C, the accuracy can achieve to 0.2°C. The imaging fiber buddle is constituted of more than 50k fibers. As the sensor probe is very thin (around 4 mm in diameter), it can also be assembled inside the radiofrequency operation knife. With the presented sensor system in clinic operation physicians can check the temperature in the operation point and view the operation area at the same time.