In this paper, the influences for one LWIR thermal camera in different integrate time are studied by using surrounding temperature simulator and blackbody of independent research and development, which compared to theoretical results. In experiment, the image of camera is obtained by using different integrate time and same temperature of blackbody at surrounding temperature of-30°C, 0°C and 20°C, the result show that the output voltage is linearly proportion to integrate time in those surrounding temperature. The experiment is studied for different blackbody temperature at surrounding temperature of-30°C, the result shows that: the slope of “integrate time vs. DL value” is increased with the target temperature. The noise of output image is increased with the increasing of integrate time. Whose agree with theoretical result. According to the conclusion, the DL value of random integrate time can be derived by known two integrate time DL for one blackbody temperature. And the calibration in random integrate time between two integrate time whose calibration curve are known, which is verified in experiment: the blackbody brightness deviation between test and derivation is less than 1%, which corresponds to blackbody temperature deviation between test and derivation is less than 1°C. The experiment results show that the measurement efficiency can be improved by using appropriate integrate time.
In the practical application of the laser, it is necessary to use the laser beam shaping technology to shape the output beam of laser device to the uniform light intensity distribution. The shaping divergent optical system of compound eye integrator way is composed of beam expanding mirror group and lens array. Its working principle is to expand the output laser to a certain size of caliber, and then divide the beam with lens array into multiple sub beam, where the lens unit of lens array can control the divergence angle of sub beam through the design of focal length, with mutual superposition of the sub beam in far field, to make up for the nonuniformity of beam, so that the radiant exitance on the radiated surface may become uniform. In this paper, we use a reflective microlens array to realize the laser beam shaping. By through of the practical optical path model established, the ray tracing is carried out and the simulation results for single-mode Gaussian beam with noise circumstance is provided. The analysis results show that the laser beam shaping under different inputs can be effectively realized by use of microlens array. All the energy is within the signal window, with a high energy efficiency of more than 90%; The measured surface has a better uniformity, and the uniformity is better than 99.5% at 150m.